CHAracterisation MEthodology Ontology
type: Ontology
version IRI: chameo
imports: mereocausality
imports: computerscience
imports: isq
imports: manufacturing
imports: math
imports: metrology
imports: models
imports: holistic
imports: persistence
imports: semiotics
imports: data
imports: persholistic
imports: workflow
abstract: CHAMEO is a domain ontology designed to model the common aspects across the different characterisation techniques and methodologies.
alternative: CHAMEO
bibliographic citation: Del Nostro, P., Goldbeck, G., Toti, D., 2022. CHAMEO: An ontology for the harmonisation of materials characterisation methodologies. Applied Ontology 17, 401–421. doi:10.3233/AO-220271.
contributor: https://orcid.org/0000-0002-4181-2852
contributor: https://orcid.org/0000-0002-5174-8508
contributor: https://orcid.org/0000-0002-9668-6961
created: 2021-12-20
creator: https://orcid.org/0000-0002-4181-2852
creator: https://orcid.org/0000-0002-5174-8508
creator: https://orcid.org/0000-0002-9668-6961
description: Characterisation Methodology Ontology
has format: turtle
issued: 2024-04-12
license: https://creativecommons.org/licenses/by/4.0/legalcode
modified: 2025-11-11
publisher: EMMC ASBL
preferred namespace prefix: chameo
comment: Contacts: Gerhard Goldbeck Goldbeck Consulting Ltd (UK) email: gerhard@goldbeck-consulting.com
prior version: 1.0.0
version info: 1.0.0
hidden label: CHAMEO

hasAccessConditions
type: ObjectProperty
domain: CharacterisationTechnique
range: AccessConditions

hasBPMNDiagram
type: ObjectProperty
range: BPMNDiagram

hasBeginCharacterisationTask
type: ObjectProperty
domain: CharacterisationWorkflow
range: CharacterisationTask
alternative label: hasBeginCharacterizationTask

hasCharacterisationComponent
type: ObjectProperty
alternative label: hasCharacterizationComponent

hasCharacterisationEnvironment
type: ObjectProperty
domain: CharacterisationProcedure
range: CharacterisationEnvironment
alternative label: hasCharacterizationEnvironment

hasCharacterisationEnvironmentProperty
type: ObjectProperty
domain: CharacterisationEnvironment
range: CharacterisationEnvironmentProperty
alternative label: hasCharacterizationEnvironmentProperty

hasCharacterisationInput
type: ObjectProperty
domain: CharacterisationProcedure
range: CharacterisationData
alternative label: hasCharacterizationInput

hasCharacterisationMeasurementInstrument
type: ObjectProperty
domain: CharacterisationMeasurementProcess
range: CharacterisationMeasurementInstrument
alternative label: hasCharacterizationMeasurementInstrument

hasCharacterisationOutput
type: ObjectProperty
domain: CharacterisationProcedure
range: CharacterisationData
alternative label: hasCharacterizationOutput

hasCharacterisationProcedureValidation
type: ObjectProperty
domain: CharacterisationTechnique
range: CharacterisationProcedureValidation

hasCharacterisationProperty
type: ObjectProperty
domain: Sample
range: CharacterisationProperty
alternative label: hasCharacterizationProperty

hasCharacterisationSoftware
type: ObjectProperty
range: CharacterisationSoftware
alternative label: hasCharacterizationSoftware

hasCharacterisationTask
type: ObjectProperty
domain: CharacterisationWorkflow
range: CharacterisationTask
alternative label: hasCharacterizationTask

hasDataAcquisitionRate
type: ObjectProperty
domain: RawData
range: DataAcquisitionRate

hasDataProcessingThroughCalibration
type: ObjectProperty
domain: CharacterisationMeasurementProcess

hasDataQuality
type: ObjectProperty
domain: DataPostProcessing
range: DataQuality

hasDataset
type: ObjectProperty

hasEndCharacterisationTask
type: ObjectProperty
domain: CharacterisationWorkflow
range: CharacterisationTask
alternative label: hasEndCharacterizationTask

hasHardwareSpecification
type: ObjectProperty
domain: CharacterisationHardware
range: CharacterisationHardwareSpecification

hasHazard
type: ObjectProperty
range: Hazard

hasHolder
type: ObjectProperty
domain: SamplePreparation
range: Holder

hasInstrumentForCalibration
type: ObjectProperty
domain: CalibrationProcess
range: CharacterisationMeasurementInstrument

hasInteractionVolume
type: ObjectProperty
domain: ProbeSampleInteraction
range: InteractionVolume

hasInteractionWithProbe
type: ObjectProperty
domain: ProbeSampleInteraction
range: Probe

hasInteractionWithSample
type: ObjectProperty
domain: ProbeSampleInteraction
range: Sample

hasLaboratory
type: ObjectProperty
range: Laboratory
alternative label: hasLab

hasLevelOfAutomation
type: ObjectProperty
domain: CharacterisationTechnique
range: LevelOfAutomation

hasMeasurementDetector
type: ObjectProperty
range: Detector

hasMeasurementParameter
type: ObjectProperty
domain: CharacterisationProcedure
range: MeasurementParameter

hasMeasurementProbe
type: ObjectProperty
range: Probe

hasMeasurementSample
type: ObjectProperty
domain: CharacterisationMeasurementProcess
range: Sample

hasMeasurementTime
type: ObjectProperty
domain: CharacterisationMeasurementProcess
range: MeasurementTime

hasOperator
type: ObjectProperty
range: Operator

hasPeerReviewedArticle
type: ObjectProperty
domain: CharacterisationProcedureValidation
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A reference to academic literature proving the scientific validity of the approach.

hasPhysicsOfInteraction
type: ObjectProperty
domain: ProbeSampleInteraction

hasPostProcessingModel
type: ObjectProperty
domain: DataPostProcessing
range: PostProcessingModel

hasProcessingReproducibility
type: ObjectProperty
domain: DataPostProcessing

hasReferenceSample
type: ObjectProperty
domain: CalibrationProcess
range: ReferenceSample

hasSampleForInspection
type: ObjectProperty
domain: SampleInspection
range: Sample

hasSampleForPreparation
type: ObjectProperty
domain: SamplePreparation
range: Sample
preferred label: hasSampleBeforeSamplePreparation

hasSampleInspectionInstrument
type: ObjectProperty
domain: SampleInspection

hasSampleInspectionParameter
type: ObjectProperty
domain: SampleInspection
range: SampleInspectionParameter

hasSamplePreparationInstrument
type: ObjectProperty
domain: SamplePreparation
range: SamplePreparationInstrument

hasSamplePreparationParameter
type: ObjectProperty
domain: SamplePreparation
range: SamplePreparationParameter

hasSampledSample
type: ObjectProperty
domain: SampleExtraction
range: Sample

rationaleHasCharacterisationProcedure
type: ObjectProperty
sub property of: topObjectProperty
domain: Rationale
range: CharacterisationProcedure

rationaleHasUserCase
type: ObjectProperty
domain: Rationale
range: UserCase

requiresLevelOfExpertise
type: ObjectProperty
range: LevelOfExpertise

hasManufacturer
type: DatatypeProperty
sub property of: topDataProperty
domain: CharacterisationHardware
range: string
comment: A string representing the Manufacturer of a CharacterisationHardware
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A string representing the Manufacturer of a CharacterisationHardware

hasModel
type: DatatypeProperty
sub property of: topDataProperty
domain: CharacterisationHardware
range: string
comment: A string representing the model of a CharacterisationHardware
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A string representing the model of a CharacterisationHardware

hasUniqueID
type: DatatypeProperty
sub property of: topDataProperty
domain: CharacterisationHardware
range: string
comment: A string representing the UniqueID of a CharacterisationHardware
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A string representing the UniqueID of a CharacterisationHardware

hasDateOfCalibration
type: DatatypeProperty
sub property of: topDataProperty
domain: CharacterisationMeasurementInstrument
range: dateTime

ACVoltammetry
type: Class
subclass of: Voltammetry
comment: Voltammetry in which a sinusoidal alternating potential of small amplitude (10 to 50 mV) of constant frequency (10 Hz to 100 kHz) is superimposed on a slowly and linearly varying potential ramp. The resulting alternating current is plotted versus imposed DC potential. The obtained AC voltammogram is peak-shaped.
alternative label: ACV
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q120895154
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Voltammetry in which a sinusoidal alternating potential of small amplitude (10 to 50 mV) of constant frequency (10 Hz to 100 kHz) is superimposed on a slowly and linearly varying potential ramp. The obtained AC voltammogram is peak-shaped.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

AbrasiveStrippingVoltammetry
type: Class
subclass of: Voltammetry
comment: Electrochemical method where traces of solid particles are abrasively transferred onto the surface of an electrode, followed by an electrochemical dissolution (anodic or cathodic dissolution) that is recorded as a current–voltage curve.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Electrochemical method where traces of solid particles are abrasively transferred onto the surface of an electrode, followed by an electrochemical dissolution (anodic or cathodic dissolution) that is recorded as a current–voltage curve.

Axiom
annotated source: AbrasiveStrippingVoltammetry
annotated target: Electrochemical method where traces of solid particles are abrasively transferred onto the surface of an electrode, followed by an electrochemical dissolution (anodic or cathodic dissolution) that is recorded as a current–voltage curve.
source: Scholz F, Nitschke L, Henrion G (1989) Naturwiss 76:71;

AccessConditions
type: Class
comment: Describes what is needed to repeat the experiment.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Describes what is needed to repeat the experiment.

AdsorptiveStrippingVoltammetry
type: Class
subclass of: StrippingVoltammetry
comment: Stripping voltammetry involving pre-concentration by adsorption of the analyte (in contrast to electro-chemical accumulation). A peak-shaped adsorptive stripping voltammogram is obtained. Peak current depends on time of accumulation, mass transport of analyte (stirring), scan rate and mode (linear or pulse), and analyte concentration in solution. AdSV is usually employed for analysis of organic compounds or metal complexes with organic ligands. Stripping is done by means of an anodic or a cathodic voltammetric scan (linear or pulse), during which the adsorbed compound is oxidized or reduced.
alternative label: AdSV
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Stripping voltammetry involving pre-concentration by adsorption of the analyte (in contrast to electro-chemical accumulation). A peak-shaped adsorptive stripping voltammogram is obtained. Peak current depends on time of accumulation, mass transport of analyte (stirring), scan rate and mode (linear or pulse), and analyte concentration in solution. AdSV is usually employed for analysis of organic compounds or metal complexes with organic ligands. Stripping is done by means of an anodic or a cathodic voltammetric scan (linear or pulse), during which the adsorbed compound is oxidized or reduced.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

AlphaSpectrometry
type: Class
subclass of: Spectrometry
comment: Alpha spectrometry (also known as alpha(-particle) spectroscopy) is the quantitative study of the energy of alpha particles emitted by a radioactive nuclide that is an alpha emitter. As emitted alpha particles are mono-energetic (i.e. not emitted with a spectrum of energies, such as beta decay) with energies often distinct to the decay they can be used to identify which radionuclide they originated from.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Alpha spectrometry (also known as alpha(-particle) spectroscopy) is the quantitative study of the energy of alpha particles emitted by a radioactive nuclide that is an alpha emitter. As emitted alpha particles are mono-energetic (i.e. not emitted with a spectrum of energies, such as beta decay) with energies often distinct to the decay they can be used to identify which radionuclide they originated from.

Amperometry
type: Class
subclass of: ElectrochemicalTesting
comment: Amperometry can be distinguished from voltammetry by the parameter being controlled (electrode potential E) and the parameter being measured (electrode current I which is usually a function of time – see chronoamperometry). In a non-stirred solution, a diffusion-limited current is usually measured, which is propor-tional to the concentration of an electroactive analyte. The current is usually faradaic and the applied potential is usually constant. The integral of current with time is the electric charge, which may be related to the amount of substance reacted by Faraday’s laws of electrolysis. The amperometric method provides the ability to distinguish selectively between a number of electroactive species in solution by judicious selection of the applied potential and/or choice of electrode material.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Amperometry can be distinguished from voltammetry by the parameter being controlled (electrode potential E) and the parameter being measured (electrode current I which is usually a function of time – see chronoamperometry). In a non-stirred solution, a diffusion-limited current is usually measured, which is propor-tional to the concentration of an electroactive analyte. The current is usually faradaic and the applied potential is usually constant. The integral of current with time is the electric charge, which may be related to the amount of substance reacted by Faraday’s laws of electrolysis. The amperometric method provides the ability to distinguish selectively between a number of electroactive species in solution by judicious selection of the applied potential and/or choice of electrode material.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

AnalyticalElectronMicroscopy
type: Class
subclass of: Microscopy
comment: Analytical electron microscopy (AEM) refers to the collection of spectroscopic data in TEM or STEM, enabling qualitative or quantitative compositional analysis.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Analytical electron microscopy (AEM) refers to the collection of spectroscopic data in TEM or STEM, enabling qualitative or quantitative compositional analysis.

AnodicStrippingVoltammetry
type: Class
subclass of: StrippingVoltammetry
comment: Stripping voltammetry in which material accumulated at the working electrode is electrochemically oxi- dized in the stripping step. A peak-shaped anodic stripping voltammogram is obtained. Peak current depends on time of accumulation, mass transport of analyte (stirring), scan rate and mode (linear or pulse), and analyte concentration in solution. A solid electrode, carbon paste or composite electrode, bismuth film electrode, mercury film electrode, or static mercury drop electrode may be used.
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q939328
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Stripping voltammetry in which material accumulated at the working electrode is electrochemically oxi- dized in the stripping step. A peak-shaped anodic stripping voltammogram is obtained. Peak current depends on time of accumulation, mass transport of analyte (stirring), scan rate and mode (linear or pulse), and analyte concentration in solution. A solid electrode, carbon paste or composite electrode, bismuth film electrode, mercury film electrode, or static mercury drop electrode may be used.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

AtomProbeTomography
type: Class
subclass of: Tomography
comment: Atom Probe Tomography (APT or 3D Atom Probe) is the only material analysis technique offering extensive capabilities for both 3D imaging and chemical composition measurements at the atomic scale (around 0.1-0.3nm resolution in depth and 0.3-0.5nm laterally). Since its early developments, Atom Probe Tomography has contributed to major advances in materials science. The sample is prepared in the form of a very sharp tip. The cooled tip is biased at high DC voltage (3-15 kV). The very small radius of the tip and the High Voltage induce a very high electrostatic field (tens V/nm) at the tip surface, just below the point of atom evaporation. Under laser or HV pulsing, one or more atoms are evaporated from the surface, by field effect (near 100% ionization), and projected onto a Position Sensitive Detector (PSD) with a very high detection efficiency. Ion efficiencies are as high as 80%, the highest analytical efficiency of any 3D microscopy.
alternative label: 3D Atom Probe
alternative label: APT
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Atom Probe Tomography (APT or 3D Atom Probe) is the only material analysis technique offering extensive capabilities for both 3D imaging and chemical composition measurements at the atomic scale (around 0.1-0.3nm resolution in depth and 0.3-0.5nm laterally). Since its early developments, Atom Probe Tomography has contributed to major advances in materials science. The sample is prepared in the form of a very sharp tip. The cooled tip is biased at high DC voltage (3-15 kV). The very small radius of the tip and the High Voltage induce a very high electrostatic field (tens V/nm) at the tip surface, just below the point of atom evaporation. Under laser or HV pulsing, one or more atoms are evaporated from the surface, by field effect (near 100% ionization), and projected onto a Position Sensitive Detector (PSD) with a very high detection efficiency. Ion efficiencies are as high as 80%, the highest analytical efficiency of any 3D microscopy.

AtomicForceMicroscopy
type: Class
subclass of: Microscopy
comment: Atomic force microscopy (AFM) is an influential surface analysis technique used for micro/nanostructured coatings. This flexible technique can be used to obtain high-resolution nanoscale images and study local sites in air (conventional AFM) or liquid (electrochemical AFM) surroundings.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Atomic force microscopy (AFM) is an influential surface analysis technique used for micro/nanostructured coatings. This flexible technique can be used to obtain high-resolution nanoscale images and study local sites in air (conventional AFM) or liquid (electrochemical AFM) surroundings.

BPMNDiagram
type: Class

BrunauerEmmettTellerMethod
type: Class
subclass of: GasAdsorptionPorosimetry
comment: A technique used to measure the specific surface area of porous materials by analyzing the adsorption of gas molecules onto the material's surface
alternative label: BET
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q795838
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A technique used to measure the specific surface area of porous materials by analyzing the adsorption of gas molecules onto the material's surface
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/BET_theory

CalibrationData
type: Class
subclass of: CharacterisationData
comment: Calibration data are used to provide correction of measured data or perform uncertainty calculations. They are generally the result of a measuerement on a reference specimen.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Calibration data are used to provide correction of measured data or perform uncertainty calculations. They are generally the result of a measuerement on a reference specimen.

CalibrationProcess
type: Class
subclass of: CharacterisationProcedure
subclass of: Restriction
  some values from: ReferenceSample
subclass of: Restriction
  some values from: MeasurementParameter
subclass of: Restriction
  some values from: CalibrationData
subclass of: Restriction
  qualified cardinality: 1
  on class: CharacterisationMeasurementInstrument
comment: Sequence of operations/actions that are needed to convert the initial signal (as produced by the detector) into a meaningful and useable raw data. From the International Vocabulary of Metrology (VIM): Operation performed on a measuring instrument or a measuring system that, under specified conditions: 1) establishes a relation between the values with measurement uncertainties provided by measurement standards and corresponding indications with associated measurement uncertainties and 2) uses this information to establish a relation for obtaining a measurement result from an indication. NOTE 1 The objective of calibration is to provide traceability of measurement results obtained when using a calibrated measuring instrument or measuring system. NOTE 2 The outcome of a calibration may be expressed by a statement, calibration function, calibration diagram, calibration curve, or calibration table. In some cases, it may consist of an additive or multiplicative correction of the indication with associated measurement uncertainty. NOTE 3 Calibration should not be confused with adjustment of a measuring system, often mistakenly called “selfcalibration”, nor with verification of calibration. Calibration is sometimes a prerequisite for verification, which provides confirmation that specified requirements (often maximum permissible errors) are met. Calibration is sometimes also a prerequisite for adjustment, which is the set of operations carried out on a measuring system such that the system provides prescribed indications corresponding to given values of quantities being measured, typically obtained from measurement standards. NOTE 4 Sometimes the first step alone of the operation mentioned in the definition is intended as being calibration, as it was in previous editions of this Vocabulary. The second step is in fact required to establish instrumental uncertainty for the measurement results obtained when using the calibrated measuring system. The two steps together aim to d…
EMMO 70fe84ff 99b6 4206 a9fc 9a8931836d84: From the International Vocabulary of Metrology (VIM): Operation performed on a measuring instrument or a measuring system that, under specified conditions: 1) establishes a relation between the values with measurement uncertainties provided by measurement standards and corresponding indications with associated measurement uncertainties and 2) uses this information to establish a relation for obtaining a measurement result from an indication. NOTE 1 The objective of calibration is to provide traceability of measurement results obtained when using a calibrated measuring instrument or measuring system. NOTE 2 The outcome of a calibration may be expressed by a statement, calibration function, calibration diagram, calibration curve, or calibration table. In some cases, it may consist of an additive or multiplicative correction of the indication with associated measurement uncertainty. NOTE 3 Calibration should not be confused with adjustment of a measuring system, often mistakenly called “selfcalibration”, nor with verification of calibration. Calibration is sometimes a prerequisite for verification, which provides confirmation that specified requirements (often maximum permissible errors) are met. Calibration is sometimes also a prerequisite for adjustment, which is the set of operations carried out on a measuring system such that the system provides prescribed indications corresponding to given values of quantities being measured, typically obtained from measurement standards. NOTE 4 Sometimes the first step alone of the operation mentioned in the definition is intended as being calibration, as it was in previous editions of this Vocabulary. The second step is in fact required to establish instrumental uncertainty for the measurement results obtained when using the calibrated measuring system. The two steps together aim to demonstrate the metrological traceability of measurement results obtained by a calibrated measuring system. In the past the second step was usually …
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Sequence of operations/actions that are needed to convert the initial signal (as produced by the detector) into a meaningful and useable raw data.
example: In nanoindentation, the electrical signal coming from capacitive displacement gauge is converted into a real raw-displacement signal after using a proper calibration function (as obtained by the equipment manufacturer). Then, additional calibration procedures are applied to define the point of initial contact and to correct for instrument compliance, thermal drift, and indenter area function to obtain the real useable displacement data.
EMMO c7b62dd7 063a 4c2a 8504 42f7264ba83f: Usually the calibration process involve a reference sample (with pre-defined, specific, and stable physical characteristics and known properties), in order to extract calibration data. In this way, the accuracy of the measurement tool and its components (for example the probe) will be evaluated and confirmed.

Calorimetry
type: Class
subclass of: ThermochemicalTesting
comment: In chemistry and thermodynamics, calorimetry (from Latin calor 'heat', and Greek μέτρον (metron) 'measure') is the science or act of measuring changes in state variables of a body for the purpose of deriving the heat transfer associated with changes of its state due, for example, to chemical reactions, physical changes, or phase transitions under specified constraints. Calorimetry is performed with a calorimeter.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: In chemistry and thermodynamics, calorimetry (from Latin calor 'heat', and Greek μέτρον (metron) 'measure') is the science or act of measuring changes in state variables of a body for the purpose of deriving the heat transfer associated with changes of its state due, for example, to chemical reactions, physical changes, or phase transitions under specified constraints. Calorimetry is performed with a calorimeter.

CathodicStrippingVoltammetry
type: Class
subclass of: StrippingVoltammetry
comment: Stripping voltammetry in which material accumulated at the working electrode is electrochemically reduced in the stripping step. A peak-shaped cathodic stripping voltammogram is obtained. Peak current depends on time of accumulation, mass transport of analyte (stirring), scan rate and mode (linear or pulse), and analyte concentration in solution.
alternative label: CSV
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q4016325
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Stripping voltammetry in which material accumulated at the working electrode is electrochemically reduced in the stripping step. A peak-shaped cathodic stripping voltammogram is obtained. Peak current depends on time of accumulation, mass transport of analyte (stirring), scan rate and mode (linear or pulse), and analyte concentration in solution.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

CharacterisationComponent
type: Class
subclass of: Restriction
  on property:
    inverse of: hasCharacterisationComponent
  some values from: CharacterisationSystem

CharacterisationData
type: Class
comment: Represents every type of data that is produced during a characterisation process
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Represents every type of data that is produced during a characterisation process

CharacterisationDataValidation
type: Class
comment: Procedure to validate the characterisation data.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Procedure to validate the characterisation data.

CharacterisationEnvironment
type: Class
subclass of: Restriction
  some values from: CharacterisationEnvironmentProperty
comment: Medium of the characterisation experiment defined by the set of environmental conditions that are controlled and measured over time during the experiment. Characterisation can either be made in air (ambient conditions, without specific controls on environmental parameters), or at different temperatures, different pressures (or in vacuum), or using different types of working gases (inert or reactive with respect to sample), different levels of humidity, etc.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Medium of the characterisation experiment defined by the set of environmental conditions that are controlled and measured over time during the experiment.
EMMO c7b62dd7 063a 4c2a 8504 42f7264ba83f: Characterisation can either be made in air (ambient conditions, without specific controls on environmental parameters), or at different temperatures, different pressures (or in vacuum), or using different types of working gases (inert or reactive with respect to sample), different levels of humidity, etc.

CharacterisationEnvironmentProperty
type: Class

CharacterisationExperiment
type: Class
comment: A characterisation experiment is the process by which a material's structure and properties are probed and measured. It is a fundamental process in the field of materials science, without which no scientific understanding of engineering materials could be ascertained.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A characterisation experiment is the process by which a material's structure and properties are probed and measured. It is a fundamental process in the field of materials science, without which no scientific understanding of engineering materials could be ascertained.

CharacterisationHardware
type: Class
comment: Whatever hardware is used during the characterisation process.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Whatever hardware is used during the characterisation process.

HardwareManufacturer
type: Class
subclass of: CharacterisationHardwareSpecification
disjoint with: HardwareModel

HardwareModel
type: Class
subclass of: CharacterisationHardwareSpecification

CharacterisationHardwareSpecification
type: Class

CharacterisationMeasurementInstrument
type: Class
subclass of: CharacterisationHardware
subclass of: Restriction
  some values from: Detector
subclass of: Restriction
  some values from: Probe
comment: The instrument used for characterising a material, which usually has a probe and a detector as parts. A device used for making measurements, alone or in conjunction with one or more supplementary devices. NOTE 1 A measuring instrument that can be used alone for making measurements is a measuring system. NOTE 2 A measuring instrument is either an indicating measuring instrument or a material measure. In nanoindentation is the nanoindenter.
EMMO 70fe84ff 99b6 4206 a9fc 9a8931836d84: A device used for making measurements, alone or in conjunction with one or more supplementary devices. NOTE 1 A measuring instrument that can be used alone for making measurements is a measuring system. NOTE 2 A measuring instrument is either an indicating measuring instrument or a material measure.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The instrument used for characterising a material, which usually has a probe and a detector as parts.
example: In nanoindentation is the nanoindenter.
EMMO bb49844b 45d7 4f0d 8cae 8e552cbc20d6: Measuring instrument

CharacterisationMeasurementProcess
type: Class
subclass of: CharacterisationProcedure
subclass of: Restriction
  some values from: MeasurementParameter
subclass of: Restriction
  some values from: CharacterisationEnvironment
subclass of: Restriction
  some values from: CharacterisationMeasurementInstrument
subclass of: Restriction
  some values from: Sample
subclass of: Restriction
  some values from: CharacterisationData
comment: The measurement process associates raw data to the sample through a probe and a detector. From the International Vocabulary of Metrology (VIM): Process of experimentally obtaining one or more values that can reasonably be attributed to a quantity together with any other available relevant information. NOTE 1 The quantity mentioned in the definition is an individual quantity. NOTE 2 The relevant information mentioned in the definition may be about the values obtained by the measurement, such that some may be more representative of the measurand than others. NOTE 3 Measurement is sometimes considered to apply to nominal properties, but not in this Vocabulary, where the process of obtaining values of nominal properties is called “examination”. NOTE 4 Measurement requires both experimental comparison of quantities or experimental counting of entities at some step of the process and the use of models and calculations that are based on conceptual considerations. NOTE 5 The conditions of reasonable attribution mentioned in the definition take into account a description of the quantity commensurate with the intended use of a measurement result, a measurement procedure, and a calibrated measuring system operating according to the specified measurement procedure, including the measurement conditions. Moreover, a maximum permissible error and/or a target uncertainty may be specified, and the measurement procedure and the measuring system should then be chosen in order not to exceed these measuring system specifications.
EMMO 70fe84ff 99b6 4206 a9fc 9a8931836d84: From the International Vocabulary of Metrology (VIM): Process of experimentally obtaining one or more values that can reasonably be attributed to a quantity together with any other available relevant information. NOTE 1 The quantity mentioned in the definition is an individual quantity. NOTE 2 The relevant information mentioned in the definition may be about the values obtained by the measurement, such that some may be more representative of the measurand than others. NOTE 3 Measurement is sometimes considered to apply to nominal properties, but not in this Vocabulary, where the process of obtaining values of nominal properties is called “examination”. NOTE 4 Measurement requires both experimental comparison of quantities or experimental counting of entities at some step of the process and the use of models and calculations that are based on conceptual considerations. NOTE 5 The conditions of reasonable attribution mentioned in the definition take into account a description of the quantity commensurate with the intended use of a measurement result, a measurement procedure, and a calibrated measuring system operating according to the specified measurement procedure, including the measurement conditions. Moreover, a maximum permissible error and/or a target uncertainty may be specified, and the measurement procedure and the measuring system should then be chosen in order not to exceed these measuring system specifications.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The measurement process associates raw data to the sample through a probe and a detector.
EMMO bb49844b 45d7 4f0d 8cae 8e552cbc20d6: Measurement

CharacterisationProcedure
type: Class
comment: The process of performing characterisation by following some existing formalised operative rules. Characterisation procedure may refer to the full characterisation process or just a part of the full process. Examples include: Sample preparation, Sample inspection, Calibration, Microscopy, Viscometry, Data sampling.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The process of performing characterisation by following some existing formalised operative rules.
example: Examples include: Sample preparation, Sample inspection, Calibration, Microscopy, Viscometry, Data sampling.
EMMO c7b62dd7 063a 4c2a 8504 42f7264ba83f: Characterisation procedure may refer to the full characterisation process or just a part of the full process.

CharacterisationProcedureValidation
type: Class
comment: Describes why the characterization procedure was chosen and deemed to be the most useful for the sample.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Describes why the characterization procedure was chosen and deemed to be the most useful for the sample.

CharacterisationProperty
type: Class
comment: The characterisation property is the investigate property or behaviour of a sample. It is derived from the secondary data, usually after classification or quantification (manually or by a model).
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The characterisation property is the investigate property or behaviour of a sample. It is derived from the secondary data, usually after classification or quantification (manually or by a model).

CharacterisationProtocol
type: Class
subclass of: CharacterisationProcedure
comment: A characterisation protocol is defined whenever it is desirable to standardize a laboratory method to ensure successful replication of results by others in the same laboratory or by other laboratories.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A characterisation protocol is defined whenever it is desirable to standardize a laboratory method to ensure successful replication of results by others in the same laboratory or by other laboratories.

CharacterisationSoftware
type: Class
comment: A software application to process characterisation data. In Nanoindentation post-processing the software used to apply the Oliver-Pharr to calculate the characterisation properties (i.e. elastic modulus, hardness) from load and depth data.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A software application to process characterisation data.
example: In Nanoindentation post-processing the software used to apply the Oliver-Pharr to calculate the characterisation properties (i.e. elastic modulus, hardness) from load and depth data.

CharacterisationSystem
type: Class
subclass of: Restriction
  some values from: CharacterisationComponent
comment: Set of one or more measuring instruments ('CharacterisationInstruments') and often other components and devices, including any sample holder, reagent and supply, assembled and adapted to give information used to generate measured values within specified intervals for quantities of specified kinds. NOTE 1 The components mentioned in the definition may be devices, reagents, and supplies. NOTE 2 A measuring system is sometimes referred to as “measuring equipment” or “device”, for example in ISO 10012, Measurement management systems – Requirements for measurement processes and measuring equipment and ISO 17025, General requirements for the competence of testing and calibration laboratories. NOTE 3 Although the terms “measuring system” and “measurement system” are frequently used synonymously, the latter is instead sometimes used to refer to a measuring system plus all other entities involved in a measurement, including the object under measurement and the person(s) performing the measurement. NOTE 4 A measuring system can be used as a measurement standard.
EMMO 70fe84ff 99b6 4206 a9fc 9a8931836d84: Set of one or more measuring instruments ('CharacterisationInstruments') and often other components and devices, including any sample holder, reagent and supply, assembled and adapted to give information used to generate measured values ('MeasuredQuantityProperty') within specified intervals for quantities of specified kinds. NOTE 1 The components mentioned in the definition may be devices, reagents, and supplies. NOTE 2 A measuring system is sometimes referred to as “measuring equipment” or “device”, for example in ISO 10012, Measurement management systems – Requirements for measurement processes and measuring equipment and ISO 17025, General requirements for the competence of testing and calibration laboratories. NOTE 3 Although the terms “measuring system” and “measurement system” are frequently used synonymously, the latter is instead sometimes used to refer to a measuring system plus all other entities involved in a measurement, including the object under measurement and the person(s) performing the measurement. NOTE 4 A measuring system can be used as a measurement standard.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Set of one or more measuring instruments ('CharacterisationInstruments') and often other components and devices, including any sample holder, reagent and supply, assembled and adapted to give information used to generate measured values ('MeasuredQuantityProperty') within specified intervals for quantities of specified kinds.
EMMO bb49844b 45d7 4f0d 8cae 8e552cbc20d6: Measuring system

CharacterisationTask
type: Class
equivalent class: Class
  intersection of: CharacterisationProcedure
subclass of: Restriction
  some values from: CharacterisationWorkflow

CharacterisationTechnique
type: Class
comment: The description of the overall characterisation technique. It can be composed of different steps (e.g. sample preparation, calibration, measurement, post-processing). A characterisation technique is not only related to the measurement process which can be one of its steps.
alternative label: Characterisation procedure
alternative label: Characterisation technique
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The description of the overall characterisation technique. It can be composed of different steps (e.g. sample preparation, calibration, measurement, post-processing).
EMMO c7b62dd7 063a 4c2a 8504 42f7264ba83f: A characterisation technique is not only related to the measurement process which can be one of its steps.

CharacterisationWorkflow
type: Class
equivalent class: Class
  intersection of: CharacterisationProcedure
subclass of: Restriction
  some values from: CharacterisationTask
comment: A characterisation procedure that has at least two characterisation tasks as proper parts.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A characterisation procedure that has at least two characterisation tasks as proper parts.

CharacterisedSample
type: Class
subclass of: Sample
comment: The sample after having been subjected to a characterization process.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The sample after having been subjected to a characterization process.

ChargeDistribution
type: Class
subclass of: CharacterisationTechnique

Chromatography
type: Class
subclass of: CharacterisationTechnique
comment: In chemical analysis, chromatography is a laboratory technique for the separation of a mixture into its components.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: In chemical analysis, chromatography is a laboratory technique for the separation of a mixture into its components.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Chromatography

Chronoamperometry
type: Class
subclass of: Amperometry
comment: Amperometry in which the current is measured as a function of time after a change in the applied potential. If the potential step is from a potential at which no current flows (i.e., at which the oxidation or reduction of the electrochemically active species does not take place) to one at which the current is limited by diffusion (see diffusion-limited current), the current obeys the Cottrell equation.
alternative label: AmperiometricDetection
alternative label: AmperometricCurrentTimeCurve
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Amperometry in which the current is measured as a function of time after a change in the applied potential. If the potential step is from a potential at which no current flows (i.e., at which the oxidation or reduction of the electrochemically active species does not take place) to one at which the current is limited by diffusion (see diffusion-limited current), the current obeys the Cottrell equation.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

Chronocoulometry
type: Class
subclass of: Coulometry
comment: Direct coulometry at controlled potential in which the electric charge passed after the application of a potential step perturbation is measured as a function of time (Q-t curve). Chronocoulometry provides the same information that is provided by chronoamperometry, since it is based on the integration of the I-t curve. Nevertheless, chronocoulometry offers important experimental advantages, such as (i) the measured signal usually increases with time and hence the later parts of the transient can be detected more accurately, (ii) a better signal-to-noise ratio can be achieved, and (iii) other contributions to overall charge passed as a function of time can be discriminated from those due to the diffusion of electroactive substances.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Direct coulometry at controlled potential in which the electric charge passed after the application of a potential step perturbation is measured as a function of time (Q-t curve). Chronocoulometry provides the same information that is provided by chronoamperometry, since it is based on the integration of the I-t curve. Nevertheless, chronocoulometry offers important experimental advantages, such as (i) the measured signal usually increases with time and hence the later parts of the transient can be detected more accurately, (ii) a better signal-to-noise ratio can be achieved, and (iii) other contributions to overall charge passed as a function of time can be discriminated from those due to the diffusion of electroactive substances.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

Chronopotentiometry
type: Class
subclass of: Potentiometry
comment: Potentiometry in which the potential is measured with time following a change in applied current. The change in applied current is usually a step, but cyclic current reversals or linearly increasing currents are also used.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Potentiometry in which the potential is measured with time following a change in applied current. The change in applied current is usually a step, but cyclic current reversals or linearly increasing currents are also used.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

CompressionTesting
type: Class
subclass of: MechanicalTesting
comment: Compression tests characterize material and product strength and stiffness under applied crushing loads. These tests are typically conducted by applying compressive pressure to a test specimen using platens or specialized fixtures with a testing machine that produces compressive loads.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Compression tests characterize material and product strength and stiffness under applied crushing loads. These tests are typically conducted by applying compressive pressure to a test specimen using platens or specialized fixtures with a testing machine that produces compressive loads.

ConductometricTitration
type: Class
subclass of: Conductometry
comment: Titration in which the electric conductivity of a solution is measured as a function of the amount of titrant added. The equivalence-point is obtained as the intersection of linear parts of the conductance G, versus titrant volume V, curve. The method can be used for deeply coloured or turbid solutions. Acid-base and precipitation reactions are most frequently used. The method is based on replacing an ionic species of the analyte with another species, cor- responding to the titrant or the product with significantly different conductance.
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q11778221
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Titration in which the electric conductivity of a solution is measured as a function of the amount of titrant added. The equivalence-point is obtained as the intersection of linear parts of the conductance G, versus titrant volume V, curve. The method can be used for deeply coloured or turbid solutions. Acid-base and precipitation reactions are most frequently used. The method is based on replacing an ionic species of the analyte with another species, cor- responding to the titrant or the product with significantly different conductance.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

Conductometry
type: Class
subclass of: ElectrochemicalTesting
comment: Measurement principle in which the electric conductivity of a solution is measured. The conductivity of a solution depends on the concentration and nature of ions present. An example is monitoring of the purity of deionized water.
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q901180
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Measurement principle in which the electric conductivity of a solution is measured. The conductivity of a solution depends on the concentration and nature of ions present.
example: Monitoring of the purity of deionized water.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Conductometry
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

ConfocalMicroscopy
type: Class
subclass of: Microscopy
comment: Confocal microscopy, most frequently confocal laser scanning microscopy (CLSM) or laser scanning confocal microscopy (LSCM), is an optical imaging technique for increasing optical resolution and contrast of a micrograph by means of using a spatial pinhole to block out-of-focus light in image formation.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Confocal microscopy, most frequently confocal laser scanning microscopy (CLSM) or laser scanning confocal microscopy (LSCM), is an optical imaging technique for increasing optical resolution and contrast of a micrograph by means of using a spatial pinhole to block out-of-focus light in image formation.

CoulometricTitration
type: Class
subclass of: Coulometry
comment: Titration in which the titrant is generated electrochemically, either by constant current or at constant potential. The titrant reacts stoichiometrically with the analyte, the amount of which is calculated using Faraday’s laws of electrolysis from the electric charge required to reach the end-point. Coulometric titrations are usually carried out in convective mass transfer mode using a large surface working electrode. The reference and auxiliary electrodes are located in sepa- rate compartments. A basic requirement is a 100 % current efficiency of titrant generation at the working electrode. End-point detection can be accomplished with potentiometry, amperometry, biamperometry, bipotentiometry, photometry, or by using a visual indicator. The main advantages are that titration is possible with less stable titrants, the standardi- zation of titrant is not necessary, the volume of the test solution is not changed, and the method is easily automated.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Titration in which the titrant is generated electrochemically, either by constant current or at constant potential. The titrant reacts stoichiometrically with the analyte, the amount of which is calculated using Faraday’s laws of electrolysis from the electric charge required to reach the end-point. Coulometric titrations are usually carried out in convective mass transfer mode using a large surface working electrode. The reference and auxiliary electrodes are located in sepa- rate compartments. A basic requirement is a 100 % current efficiency of titrant generation at the working electrode. End-point detection can be accomplished with potentiometry, amperometry, biamperometry, bipotentiometry, photometry, or by using a visual indicator. The main advantages are that titration is possible with less stable titrants, the standardi- zation of titrant is not necessary, the volume of the test solution is not changed, and the method is easily automated.

Coulometry
type: Class
subclass of: ElectrochemicalTesting
comment: Electrochemical measurement principle in which the electric charge required to carry out a known electrochemical reaction is measured. By Faraday’s laws of electrolysis, the amount of substance is proportional to the charge. Coulometry used to measure the amount of substance is a primary reference measurement procedure [VIM 2.8] not requiring calibration with a standard for a quantity of the same kind (i.e. amount of substance). The coulometric experiment can be carried out at controlled (constant) potential (see direct coulometry at controlled potential) or controlled (constant) current (see direct coulometry at controlled current).
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q1136979
EMMO 50c298c2 55a2 4068 b3ac 4e948c33181f: https://www.electropedia.org/iev/iev.nsf/display
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Electrochemical measurement principle in which the electric charge required to carry out a known electrochemical reaction is measured. By Faraday’s laws of electrolysis, the amount of substance is proportional to the charge. Coulometry used to measure the amount of substance is a primary reference measurement procedure [VIM 2.8] not requiring calibration with a standard for a quantity of the same kind (i.e. amount of substance). The coulometric experiment can be carried out at controlled (constant) potential (see direct coulometry at controlled potential) or controlled (constant) current (see direct coulometry at controlled current).
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Coulometry
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

CreepTesting
type: Class
subclass of: MechanicalTesting
comment: The creep test is a destructive materials testing method for determination of the long-term strength and heat resistance of a material. When running a creep test, the specimen is subjected to increased temperature conditions for an extended period of time and loaded with a constant tensile force or tensile stress.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The creep test is a destructive materials testing method for determination of the long-term strength and heat resistance of a material. When running a creep test, the specimen is subjected to increased temperature conditions for an extended period of time and loaded with a constant tensile force or tensile stress.

CriticalAndSupercriticalChromatography
type: Class
subclass of: Chromatography

CyclicChronopotentiometry
type: Class
subclass of: Chronopotentiometry
comment: Chronopotentiometry where the change in applied current undergoes a cyclic current reversal.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Chronopotentiometry where the change in applied current undergoes a cyclic current reversal.

Axiom
annotated source: CyclicChronopotentiometry
annotated target: Chronopotentiometry where the change in applied current undergoes a cyclic current reversal.
source: J. M. Pingarrón et al., Terminology of electrochemical methods of analysis (IUPAC Recommendations 2019), Pure and Applied Chemistry, 4, 92, 2020, 641-694. https://doi.org/10.1515/pac-2018-0109.

CyclicVoltammetry
type: Class
subclass of: Voltammetry
comment: Voltammetry in which the electric current is recorded as the electrode potential is varied with time cyclically between two potential limits, normally at a constant scan rate. Cyclic voltammetry is frequently used for the investigation of mechanisms of electrochemical/electrode reactions. The current-potential curve may be modelled to obtain reaction mechanisms and electrochemical parameters. Normally the initial potential is chosen where no electrode reaction occurs and the switching potential is greater (more positive for an oxidation or more negative for a reduction) than the peak potential of the analyte reaction. The initial potential is usually the negative or positive limit of the cycle but can have any value between the two limits, as can the initial scan direction. The limits of the potential are known as the switching potentials. The plot of current against potential is termed a cyclic voltammogram. Usually peak-shaped responses are obtained for scans in both directions.
alternative label: CV
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q1147647
EMMO 6dd685dd 1895 46e4 b227 be9f7d643c25: https://dbpedia.org/page/Cyclic_voltammetry
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Voltammetry in which the electric current is recorded as the electrode potential is varied with time cyclically between two potential limits, normally at a constant scan rate. Cyclic voltammetry is frequently used for the investigation of mechanisms of electrochemical/electrode reactions. The current-potential curve may be modelled to obtain reaction mechanisms and electrochemical parameters. Normally the initial potential is chosen where no electrode reaction occurs and the switching potential is greater (more positive for an oxidation or more negative for a reduction) than the peak potential of the analyte reaction. The initial potential is usually the negative or positive limit of the cycle but can have any value between the two limits, as can the initial scan direction. The limits of the potential are known as the switching potentials. The plot of current against potential is termed a cyclic voltammogram. Usually peak-shaped responses are obtained for scans in both directions.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Cyclic_voltammetry
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

DCPolarography
type: Class
subclass of: Voltammetry
comment: Linear scan voltammetry with slow scan rate in which a dropping mercury electrode is used as the working electrode. If the whole scan is performed on a single growing drop, the technique should be called single drop scan voltammetry. The term polarography in this context is discouraged. This is the oldest variant of polarographic techniques, introduced by Jaroslav Heyrovský (1890 – 1967). Usually the drop time is between 1 and 5 s and the pseudo-steady-state wave-shaped dependence on potential is called a polarogram. If the limiting current is controlled by diffusion, it is expressed by the Ilkovich equation.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Linear scan voltammetry with slow scan rate in which a dropping mercury electrode is used as the working electrode. If the whole scan is performed on a single growing drop, the technique should be called single drop scan voltammetry. The term polarography in this context is discouraged. This is the oldest variant of polarographic techniques, introduced by Jaroslav Heyrovský (1890 – 1967). Usually the drop time is between 1 and 5 s and the pseudo-steady-state wave-shaped dependence on potential is called a polarogram. If the limiting current is controlled by diffusion, it is expressed by the Ilkovich equation.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

DataAcquisitionRate
type: Class
comment: Quantifies the raw data acquisition rate, if applicable.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Quantifies the raw data acquisition rate, if applicable.

DataAnalysis
type: Class
comment: Data processing activities performed on the secondary data to determine the characterisation property (e.g. classification, quantification), which can be performed manually or exploiting a model.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Data processing activities performed on the secondary data to determine the characterisation property (e.g. classification, quantification), which can be performed manually or exploiting a model.

DataFiltering
type: Class
subclass of: DataPreparation
comment: Data filtering is the process of examining a dataset to exclude, rearrange, or apportion data according to certain criteria.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Data filtering is the process of examining a dataset to exclude, rearrange, or apportion data according to certain criteria.

DataNormalisation
type: Class
subclass of: DataPreparation
comment: Data normalization involves adjusting raw data to a notionally common scale.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Data normalization involves adjusting raw data to a notionally common scale.
EMMO c7b62dd7 063a 4c2a 8504 42f7264ba83f: It involves the creation of shifted and/or scaled versions of the values to allow post-processing in a way that eliminates the effects of influences on subsequent properties extraction.

DataPostProcessing
type: Class
comment: Analysis, that allows one to calculate the final material property from the calibrated primary data.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Analysis, that allows one to calculate the final material property from the calibrated primary data.

DataPreparation
type: Class
comment: Data preparation is the process of manipulating (or pre-processing) data (which may come from disparate data sources) to improve their quality or reduce bias in subsequent analysis.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Data preparation is the process of manipulating (or pre-processing) data (which may come from disparate data sources) to improve their quality or reduce bias in subsequent analysis.

DataQuality
type: Class
comment: Evaluation of quality indicators to determine how well suited a data set is to be used for the characterisation of a material. An example is the evaluation of S/N ratio, or other quality indicators (limits of detection/quantification, statistical analysis of data, data robustness analysis).
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Evaluation of quality indicators to determine how well suited a data set is to be used for the characterisation of a material.
example: An example is the evaluation of S/N ratio, or other quality indicators (limits of detection/quantification, statistical analysis of data, data robustness analysis).

Detector
type: Class
subclass of: CharacterisationHardware
comment: Physical device (or the chain of devices) that is used to measure, quantify and store the signal after its interaction with the sample. Examples include: Back Scattered Electrons (BSE) and Secondary Electrons (SE) detectors for SEM; Displacement and force sensors for mechanical testing.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Physical device (or the chain of devices) that is used to measure, quantify and store the signal after its interaction with the sample.
example: Examples include: Back Scattered Electrons (BSE) and Secondary Electrons (SE) detectors for SEM; Displacement and force sensors for mechanical testing.

DeviceSample
type: Class
subclass of: Sample
comment: Specific type of sample that is a device.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Specific type of sample that is a device.

DielectricAndImpedanceSpectroscopy
type: Class
subclass of: Spectroscopy
comment: Dielectric spectroscopy (DS) or impedance spectroscopy, also known as electrochemical impedance spectroscopy, is frequently used to study the response of a sample subjected to an applied electric field of fixed or changing frequency. DS describes the dielectric properties of a material as a function of frequency. In DS, the radio and microwave frequency regions of the electromagnetic spectrum have been successfully made to interact with materials, so as to study the behavior of molecules. The interaction of applied alternating electric fields with dipoles possessing reorientation mobility in materials is also dealt by DS.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Dielectric spectroscopy (DS) or impedance spectroscopy, also known as electrochemical impedance spectroscopy, is frequently used to study the response of a sample subjected to an applied electric field of fixed or changing frequency. DS describes the dielectric properties of a material as a function of frequency. In DS, the radio and microwave frequency regions of the electromagnetic spectrum have been successfully made to interact with materials, so as to study the behavior of molecules. The interaction of applied alternating electric fields with dipoles possessing reorientation mobility in materials is also dealt by DS.

Dielectrometry
type: Class
subclass of: ElectrochemicalTesting
comment: Electrochemical measurement principle based on the measurement of the dielectric constant of a sample resulting from the orientation of particles (molecules or ions) that have a dipole moment in an electric field. Dielectrometric titrations use dielectrometry for the end-point detection. The method is used to monitor the purity of dielectrics, for example to detect small amounts of moisture.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Electrochemical measurement principle based on the measurement of the dielectric constant of a sample resulting from the orientation of particles (molecules or ions) that have a dipole moment in an electric field. Dielectrometric titrations use dielectrometry for the end-point detection. The method is used to monitor the purity of dielectrics, for example to detect small amounts of moisture.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

DifferentialLinearPulseVoltammetry
type: Class
subclass of: DifferentialPulseVoltammetry
comment: Differential Pulse Voltammetry in which small potential pulses are superimposed onto a linearly varying potential.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Differential Pulse Voltammetry in which small potential pulses are superimposed onto a linearly varying potential.

Axiom
annotated source: DifferentialLinearPulseVoltammetry
annotated target: Differential Pulse Voltammetry in which small potential pulses are superimposed onto a linearly varying potential.
source: J. M. Pingarrón et al., Terminology of electrochemical methods of analysis (IUPAC Recommendations 2019), Pure and Applied Chemistry, 4, 92, 2020, 641-694. https://doi.org/10.1515/pac-2018-0109

DifferentialPulseVoltammetry
type: Class
subclass of: Voltammetry
comment: Voltammetry in which small potential pulses (constant height 10 to 100 mV, constant width 10 to 100 ms) are superimposed onto a linearly varying potential or onto a staircase potential ramp. The current is sampled just before the onset of the pulse (e.g. 10 to 20 ms) and for the same sampling time just before the end of the pulse. The difference between the two sampled currents is plotted versus the potential applied before the pulse. Thus, a differential pulse voltammogram is peak-shaped. Differential pulse polarography is differential pulse voltammetry in which a dropping mercury electrode is used as the working electrode. A pulse is applied before the mechani- cally enforced end of the drop and the current is sampled twice: just before the onset of the pulse and just before its end. The pulse width is usually 10 to 20 % of the drop life. The drop dislodgement is synchronized with current sampling, which is carried out as in DPV. The ratio of faradaic current to charging current is enhanced and the negative influence of charging current is partially eliminated in the same way as in normal pulse voltammetry (NPV). Moreover, subtraction of the charging current sampled before the application of the pulse further decreases its negative influence. Due to the more enhanced signal (faradaic current) to noise (charging current) ratio, the limit of detection is lower than with NPV. The sensitivity of DPV depends on the reversibility of the electrode reaction of the analyte.
alternative label: DPV
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q5275361
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Voltammetry in which small potential pulses (constant height 10 to 100 mV, constant width 10 to 100 ms) are superimposed onto a linearly varying potential or onto a staircase potential ramp. The current is sampled just before the onset of the pulse (e.g. 10 to 20 ms) and for the same sampling time just before the end of the pulse. The difference between the two sampled currents is plotted versus the potential applied before the pulse. Thus, a differential pulse voltammogram is peak-shaped. Differential pulse polarography is differential pulse voltammetry in which a dropping mercury electrode is used as the working electrode. A pulse is applied before the mechani- cally enforced end of the drop and the current is sampled twice: just before the onset of the pulse and just before its end. The pulse width is usually 10 to 20 % of the drop life. The drop dislodgement is synchronized with current sampling, which is carried out as in DPV. The ratio of faradaic current to charging current is enhanced and the negative influence of charging current is partially eliminated in the same way as in normal pulse voltammetry (NPV). Moreover, subtraction of the charging current sampled before the application of the pulse further decreases its negative influence. Due to the more enhanced signal (faradaic current) to noise (charging current) ratio, the limit of detection is lower than with NPV. The sensitivity of DPV depends on the reversibility of the electrode reaction of the analyte.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Differential_pulse_voltammetry
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

DifferentialRefractiveIndex
type: Class
subclass of: OpticalTesting

DifferentialScanningCalorimetry
type: Class
subclass of: ThermochemicalTesting
comment: Differential scanning calorimetry (DSC) is a thermoanalytical technique in which the difference in the amount of heat required to increase the temperature of a sample and reference is measured as a function of temperature. Both the sample and reference are maintained at nearly the same temperature throughout the experiment. Generally, the temperature program for a DSC analysis is designed such that the sample holder temperature increases linearly as a function of time. The reference sample should have a well-defined heat capacity over the range of temperatures to be scanned. Additionally, the reference sample must be stable, of high purity, and must not experience much change across the temperature scan. Typically, reference standards have been metals such as indium, tin, bismuth, and lead, but other standards such as polyethylene and fatty acids have been proposed to study polymers and organic compounds, respectively.
alternative label: DSC
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Differential scanning calorimetry (DSC) is a thermoanalytical technique in which the difference in the amount of heat required to increase the temperature of a sample and reference is measured as a function of temperature. Both the sample and reference are maintained at nearly the same temperature throughout the experiment. Generally, the temperature program for a DSC analysis is designed such that the sample holder temperature increases linearly as a function of time. The reference sample should have a well-defined heat capacity over the range of temperatures to be scanned. Additionally, the reference sample must be stable, of high purity, and must not experience much change across the temperature scan. Typically, reference standards have been metals such as indium, tin, bismuth, and lead, but other standards such as polyethylene and fatty acids have been proposed to study polymers and organic compounds, respectively.

DifferentialStaircasePulseVoltammetry
type: Class
subclass of: DifferentialPulseVoltammetry
comment: Differential Pulse Voltammetry in which small potential pulses are superimposed onto a staircase potential ramp.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Differential Pulse Voltammetry in which small potential pulses are superimposed onto a staircase potential ramp.

Axiom
annotated source: DifferentialStaircasePulseVoltammetry
annotated target: Differential Pulse Voltammetry in which small potential pulses are superimposed onto a staircase potential ramp.
source: J. M. Pingarrón et al., Terminology of electrochemical methods of analysis (IUPAC Recommendations 2019), Pure and Applied Chemistry, 4, 92, 2020, 641-694. https://doi.org/10.1515/pac-2018-0109

DifferentialThermalAnalysis
type: Class
subclass of: ThermochemicalTesting
comment: Differential thermal analysis (DTA) is a thermoanalytic technique that is similar to differential scanning calorimetry. In DTA, the material under study and an inert reference are made to undergo identical thermal cycles, (i.e., same cooling or heating programme) while recording any temperature difference between sample and reference.[1] This differential temperature is then plotted against time, or against temperature (DTA curve, or thermogram). Changes in the sample, either exothermic or endothermic, can be detected relative to the inert reference. Thus, a DTA curve provides data on the transformations that have occurred, such as glass transitions, crystallization, melting and sublimation. The area under a DTA peak is the enthalpy change and is not affected by the heat capacity of the sample.
alternative label: DTA
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Differential thermal analysis (DTA) is a thermoanalytic technique that is similar to differential scanning calorimetry. In DTA, the material under study and an inert reference are made to undergo identical thermal cycles, (i.e., same cooling or heating programme) while recording any temperature difference between sample and reference.[1] This differential temperature is then plotted against time, or against temperature (DTA curve, or thermogram). Changes in the sample, either exothermic or endothermic, can be detected relative to the inert reference. Thus, a DTA curve provides data on the transformations that have occurred, such as glass transitions, crystallization, melting and sublimation. The area under a DTA peak is the enthalpy change and is not affected by the heat capacity of the sample.

Dilatometry
type: Class
subclass of: CharacterisationTechnique
comment: Dilatometry is a method for characterising the dimensional changes of materials with variation of temperature conditions.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Dilatometry is a method for characterising the dimensional changes of materials with variation of temperature conditions.

DirectCoulometryAtControlledCurrent
type: Class
subclass of: Coulometry
comment: Coulometry at an imposed, constant current in the electrochemical cell. Direct coulometry at controlled current is usually carried out in convective mass transfer mode. The end-point of the electrolysis, at which the current is stopped, must be determined either from the inflection point in the E–t curve or by using visual or objective end-point indi- cation, similar to volumetric methods. The total electric charge is calculated as the product of the constant current and time of electrolysis or can be measured directly using a coulometer. The advantage of this method is that the electric charge consumed during the electrode reaction is directly proportional to the electrolysis time. Care must be taken to avoid the potential region where another electrode reaction may occur.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Coulometry at an imposed, constant current in the electrochemical cell. Direct coulometry at controlled current is usually carried out in convective mass transfer mode. The end-point of the electrolysis, at which the current is stopped, must be determined either from the inflection point in the E–t curve or by using visual or objective end-point indi- cation, similar to volumetric methods. The total electric charge is calculated as the product of the constant current and time of electrolysis or can be measured directly using a coulometer. The advantage of this method is that the electric charge consumed during the electrode reaction is directly proportional to the electrolysis time. Care must be taken to avoid the potential region where another electrode reaction may occur.

DirectCoulometryAtControlledPotential
type: Class
subclass of: Coulometry
comment: Coulometry at a preselected constant potential of the working electrode. Direct coulometry at controlled potential is usually carried out in convective mass trans- fer mode using a large surface working electrode. Reference and auxiliary electrodes are placed in separate compartments. The total electric charge is obtained by integration of the I–t curve or can be measured directly using a coulometer.
comment: In principle, the end point at which I = 0, i.e. when the concentration of species under study becomes zero, can be reached only at infinite time. However, in practice, the electrolysis is stopped when the current has decayed to a few percent of the initial value and the charge passed at infinite time is calculated from a plot of charge Q(t) against time t. For a simple system under diffusion control Qt= Q∞[1 − exp(−DAt/Vδ)], where Q∞ = limt→∞Q(t) is the total charge passed at infinite time, D is the diffusion coefficient of the electroactive species, A the electrode area, δ the diffusion layer thickness, and V the volume of the solution.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Coulometry at a preselected constant potential of the working electrode. Direct coulometry at controlled potential is usually carried out in convective mass trans- fer mode using a large surface working electrode. Reference and auxiliary electrodes are placed in separate compartments. The total electric charge is obtained by integration of the I–t curve or can be measured directly using a coulometer.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: In principle, the end point at which I = 0, i.e. when the concentration of species under study becomes zero, can be reached only at infinite time. However, in practice, the electrolysis is stopped when the current has decayed to a few percent of the initial value and the charge passed at infinite time is calculated from a plot of charge Q(t) against time t. For a simple system under diffusion control Qt= Q∞[1 − exp(−DAt/Vδ)], where Q∞ = limt→∞Q(t) is the total charge passed at infinite time, D is the diffusion coefficient of the electroactive species, A the electrode area, δ the diffusion layer thickness, and V the volume of the solution.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

DirectCurrentInternalResistance
type: Class
subclass of: Chronopotentiometry
comment: Method of determining the internal resistance of an electrochemical cell by applying a low current followed by higher current within a short period, and then record the changes of battery voltage and current.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Method of determining the internal resistance of an electrochemical cell by applying a low current followed by higher current within a short period, and then record the changes of battery voltage and current.

DynamicLightScattering
type: Class
subclass of: OpticalTesting
comment: Dynamic light scattering (DLS) is a technique in physics that can be used to determine the size distribution profile of small particles in suspension or polymers in solution. In the scope of DLS, temporal fluctuations are usually analyzed using the intensity or photon auto-correlation function (also known as photon correlation spectroscopy - PCS or quasi-elastic light scattering - QELS).
alternative label: DLS
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Dynamic light scattering (DLS) is a technique in physics that can be used to determine the size distribution profile of small particles in suspension or polymers in solution. In the scope of DLS, temporal fluctuations are usually analyzed using the intensity or photon auto-correlation function (also known as photon correlation spectroscopy - PCS or quasi-elastic light scattering - QELS).

DynamicMechanicalAnalysis
type: Class
subclass of: MechanicalTesting
comment: Dynamic mechanical analysis (abbreviated DMA) is a characterisation technique where a sinusoidal stress is applied and the strain in the material is measured, allowing one to determine the complex modulus. The temperature of the sample or the frequency of the stress are often varied, leading to variations in the complex modulus; this approach can be used to locate the glass transition temperature[1] of the material, as well as to identify transitions corresponding to other molecular motions.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Dynamic mechanical analysis (abbreviated DMA) is a characterisation technique where a sinusoidal stress is applied and the strain in the material is measured, allowing one to determine the complex modulus. The temperature of the sample or the frequency of the stress are often varied, leading to variations in the complex modulus; this approach can be used to locate the glass transition temperature[1] of the material, as well as to identify transitions corresponding to other molecular motions.

DynamicMechanicalSpectroscopy
type: Class
subclass of: Spectroscopy
comment: Dynamic Mechanical Analysis (DMA) is a material characterization technique where a small deformation is applied to a sample in a cyclic manner. This allows measurement of the materials response to stress, temperature, frequency or time. The term is also used to refer to the analyzer that performs the test.
alternative label: DMA
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Dynamic Mechanical Analysis (DMA) is a material characterization technique where a small deformation is applied to a sample in a cyclic manner. This allows measurement of the materials response to stress, temperature, frequency or time. The term is also used to refer to the analyzer that performs the test.

ElectrochemicalImpedanceSpectroscopy
type: Class
subclass of: Impedimetry
comment: Electrochemical measurement method of the complex impedance of an electrochemical system as a function of the frequency of a small amplitude (normally 5 to 10 mV) sinusoidal voltage perturbation superimposed on a fixed value of applied potential or on the open circuit potential. Impedimetric sensors are based on measurement of a concentration-dependent parameter taken from analysis of the respective electrochemical impedance spectra, or from the impedance magnitudes at a chosen fixed frequency. The sinusoidal current response lags behind the sinusoidal voltage perturbation by a phase angle φ. Resistances (e.g. to charge transfer) give a response in phase with the voltage perturbation; capacitances (e.g. double layer) give a response 90° out of phase; combinations of resistances and capacitances give phase angles between 0 and 90°. Plots of the out of phase vs. the in phase component of the impedance for all the frequencies tested are called complex plane (or Nyquist) plots. Plots of the phase angle and the magnitude of the impedance vs. the logarithm of perturbation frequency are called Bode diagrams. Complex plane plots are the more commonly used for electrochemical sensors.
alternative label: EIS
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q3492904
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Electrochemical measurement method of the complex impedance of an electrochemical system as a function of the frequency of a small amplitude (normally 5 to 10 mV) sinusoidal voltage perturbation superimposed on a fixed value of applied potential or on the open circuit potential. Impedimetric sensors are based on measurement of a concentration-dependent parameter taken from analysis of the respective electrochemical impedance spectra, or from the impedance magnitudes at a chosen fixed frequency. The sinusoidal current response lags behind the sinusoidal voltage perturbation by a phase angle φ. Resistances (e.g. to charge transfer) give a response in phase with the voltage perturbation; capacitances (e.g. double layer) give a response 90° out of phase; combinations of resistances and capacitances give phase angles between 0 and 90°. Plots of the out of phase vs. the in phase component of the impedance for all the frequencies tested are called complex plane (or Nyquist) plots. Plots of the phase angle and the magnitude of the impedance vs. the logarithm of perturbation frequency are called Bode diagrams. Complex plane plots are the more commonly used for electrochemical sensors.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

ElectrochemicalPiezoelectricMicrogravimetry
type: Class
subclass of: Electrogravimetry
comment: Electrogravimetry using an electrochemical quartz crystal microbalance. The change of mass is, for rigid deposits, linearly proportional to the change of the reso- nance frequency of the quartz crystal, according to the Sauerbrey equation. For non- rigid deposits, corrections must be made.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Electrogravimetry using an electrochemical quartz crystal microbalance. The change of mass is, for rigid deposits, linearly proportional to the change of the reso- nance frequency of the quartz crystal, according to the Sauerbrey equation. For non- rigid deposits, corrections must be made.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

ElectrochemicalTesting
type: Class
subclass of: ChargeDistribution
comment: In electrochemical characterization, the measurement of potential, charge, or current is used to determine an analyte's concentration or to characterize an analyte's chemical reactivity
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: In electrochemical characterization, the measurement of potential, charge, or current is used to determine an analyte's concentration or to characterize an analyte's chemical reactivity.

Electrogravimetry
type: Class
subclass of: ElectrochemicalTesting
comment: Method of electroanalytical chemistry used to separate by electrolyse ions of a substance and to derive the amount of this substance from the increase in mass of an electrode.
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q902953
EMMO 50c298c2 55a2 4068 b3ac 4e948c33181f: https://www.electropedia.org/iev/iev.nsf/display
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Method of electroanalytical chemistry used to separate by electrolyse ions of a substance and to derive the amount of this substance from the increase in mass of an electrode.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: method of electroanalytical chemistry used to separate by electrolyse ions of a substance and to derive the amount of this substance from the increase in mass of an electrode.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Electrogravimetry

Axiom
annotated source: Electrogravimetry
annotated target: method of electroanalytical chemistry used to separate by electrolyse ions of a substance and to derive the amount of this substance from the increase in mass of an electrode.
source: International Electrotechnical Commission (IEC), IEC 60050 - International Electrotechnical Vocabulary, retrieved from: https://www.electropedia.org

ElectronBackscatterDiffraction
type: Class
subclass of: ScanningElectronMicroscopy
subclass of: ScatteringAndDiffraction
comment: Electron backscatter diffraction (EBSD) is a scanning electron microscopy (SEM) technique used to study the crystallographic structure of materials. EBSD is carried out in a scanning electron microscope equipped with an EBSD detector comprising at least a phosphorescent screen, a compact lens and a low-light camera. In this configuration, the SEM incident beam hits the tilted sample. As backscattered electrons leave the sample, they interact with the crystal's periodic atomic lattice planes and diffract according to Bragg's law at various scattering angles before reaching the phosphor screen forming Kikuchi patterns (EBSPs). EBSD spatial resolution depends on many factors, including the nature of the material under study and the sample preparation. Thus, EBSPs can be indexed to provide information about the material's grain structure, grain orientation, and phase at the micro-scale. EBSD is applied for impurities and defect studies, plastic deformation, and statistical analysis for average misorientation, grain size, and crystallographic texture. EBSD can also be combined with energy-dispersive X-ray spectroscopy (EDS), cathodoluminescence (CL), and wavelength-dispersive X-ray spectroscopy (WDS) for advanced phase identification and materials discovery.
alternative label: EBSD
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Electron backscatter diffraction (EBSD) is a scanning electron microscopy (SEM) technique used to study the crystallographic structure of materials. EBSD is carried out in a scanning electron microscope equipped with an EBSD detector comprising at least a phosphorescent screen, a compact lens and a low-light camera. In this configuration, the SEM incident beam hits the tilted sample. As backscattered electrons leave the sample, they interact with the crystal's periodic atomic lattice planes and diffract according to Bragg's law at various scattering angles before reaching the phosphor screen forming Kikuchi patterns (EBSPs). EBSD spatial resolution depends on many factors, including the nature of the material under study and the sample preparation. Thus, EBSPs can be indexed to provide information about the material's grain structure, grain orientation, and phase at the micro-scale. EBSD is applied for impurities and defect studies, plastic deformation, and statistical analysis for average misorientation, grain size, and crystallographic texture. EBSD can also be combined with energy-dispersive X-ray spectroscopy (EDS), cathodoluminescence (CL), and wavelength-dispersive X-ray spectroscopy (WDS) for advanced phase identification and materials discovery.

ElectronProbeMicroanalysis
type: Class
subclass of: Microscopy
comment: Electron probe microanalysis (EPMA) is used for quantitative analysis of the elemental composition of solid specimens at a micrometer scale. The method uses bombardment of the specimen by keV electrons to excite characteristic X-rays from the sample, which are then detected by using wavelength-dispersive (WD) spectrometers.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Electron probe microanalysis (EPMA) is used for quantitative analysis of the elemental composition of solid specimens at a micrometer scale. The method uses bombardment of the specimen by keV electrons to excite characteristic X-rays from the sample, which are then detected by using wavelength-dispersive (WD) spectrometers.

Ellipsometry
type: Class
subclass of: OpticalTesting
comment: Ellipsometry is an optical technique that uses polarised light to probe the dielectric properties of a sample (optical system). The common application of ellipsometry is the analysis of thin films. Through the analysis of the state of polarisation of the light that is reflected from the sample, ellipsometry yields information on the layers that are thinner than the wavelength of the light itself, down to a single atomic layer or less. Depending on what is already known about the sample, the technique can probe a range of properties including layer thickness, morphology, and chemical composition.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Ellipsometry is an optical technique that uses polarised light to probe the dielectric properties of a sample (optical system). The common application of ellipsometry is the analysis of thin films. Through the analysis of the state of polarisation of the light that is reflected from the sample, ellipsometry yields information on the layers that are thinner than the wavelength of the light itself, down to a single atomic layer or less. Depending on what is already known about the sample, the technique can probe a range of properties including layer thickness, morphology, and chemical composition.

EnergyDispersiveXraySpectroscopy
type: Class
subclass of: Spectroscopy
comment: An analytical technique used for the elemental analysis or chemical characterization of a sample.
alternative label: EDS
alternative label: EDX
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q386334
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: An analytical technique used for the elemental analysis or chemical characterization of a sample.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Energy-dispersive_X-ray_spectroscopy

EnvironmentalScanningElectronMicroscopy
type: Class
subclass of: Microscopy
comment: The environmental scanning electron microscope (ESEM) is a scanning electron microscope (SEM) that allows for the option of collecting electron micrographs of specimens that are wet, uncoated, or both by allowing for a gaseous environment in the specimen chamber.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The environmental scanning electron microscope (ESEM) is a scanning electron microscope (SEM) that allows for the option of collecting electron micrographs of specimens that are wet, uncoated, or both by allowing for a gaseous environment in the specimen chamber.

Exafs
type: Class
subclass of: Spectroscopy
comment: Extended X-ray absorption fine structure (EXAFS), along with X-ray absorption near edge structure (XANES), is a subset of X-ray absorption spectroscopy (XAS). Like other absorption spectroscopies, XAS techniques follow Beer's law. The X-ray absorption coefficient of a material as a function of energy is obtained by directing X-rays of a narrow energy range at a sample, while recording the incident and transmitted x-ray intensity, as the incident x-ray energy is incremented. When the incident x-ray energy matches the binding energy of an electron of an atom within the sample, the number of x-rays absorbed by the sample increases dramatically, causing a drop in the transmitted x-ray intensity. This results in an absorption edge. Every element has a set of unique absorption edges corresponding to different binding energies of its electrons, giving XAS element selectivity. XAS spectra are most often collected at synchrotrons because of the high intensity of synchrotron X-ray sources allow the concentration of the absorbing element to reach as low as a few parts per million. Absorption would be undetectable if the source is too weak. Because X-rays are highly penetrating, XAS samples can be gases, solids or liquids.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Extended X-ray absorption fine structure (EXAFS), along with X-ray absorption near edge structure (XANES), is a subset of X-ray absorption spectroscopy (XAS). Like other absorption spectroscopies, XAS techniques follow Beer's law. The X-ray absorption coefficient of a material as a function of energy is obtained by directing X-rays of a narrow energy range at a sample, while recording the incident and transmitted x-ray intensity, as the incident x-ray energy is incremented. When the incident x-ray energy matches the binding energy of an electron of an atom within the sample, the number of x-rays absorbed by the sample increases dramatically, causing a drop in the transmitted x-ray intensity. This results in an absorption edge. Every element has a set of unique absorption edges corresponding to different binding energies of its electrons, giving XAS element selectivity. XAS spectra are most often collected at synchrotrons because of the high intensity of synchrotron X-ray sources allow the concentration of the absorbing element to reach as low as a few parts per million. Absorption would be undetectable if the source is too weak. Because X-rays are highly penetrating, XAS samples can be gases, solids or liquids.

FatigueTesting
type: Class
subclass of: MechanicalTesting
comment: Fatigue testing is a specialised form of mechanical testing that is performed by applying cyclic loading to a coupon or structure. These tests are used either to generate fatigue life and crack growth data, identify critical locations or demonstrate the safety of a structure that may be susceptible to fatigue.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Fatigue testing is a specialised form of mechanical testing that is performed by applying cyclic loading to a coupon or structure. These tests are used either to generate fatigue life and crack growth data, identify critical locations or demonstrate the safety of a structure that may be susceptible to fatigue.

FibDic
type: Class
subclass of: MechanicalTesting
comment: The FIB-DIC (Focused Ion Beam - Digital Image Correlation) ring-core technique is a powerful method for measuring residual stresses in materials. It is based on milling a ring-shaped sample, or core, from the material of interest using a focused ion beam (FIB).
alternative label: FIBDICResidualStressAnalysis
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The FIB-DIC (Focused Ion Beam - Digital Image Correlation) ring-core technique is a powerful method for measuring residual stresses in materials. It is based on milling a ring-shaped sample, or core, from the material of interest using a focused ion beam (FIB).

FieldEmissionScanningElectronMicroscopy
type: Class
subclass of: Microscopy
comment: Field emission scanning electron microscopy (FE-SEM) is an advanced technology used to capture the microstructure image of the materials. FE-SEM is typically performed in a high vacuum because gas molecules tend to disturb the electron beam and the emitted secondary and backscattered electrons used for imaging.
alternative label: FE-SEM
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Field emission scanning electron microscopy (FE-SEM) is an advanced technology used to capture the microstructure image of the materials. FE-SEM is typically performed in a high vacuum because gas molecules tend to disturb the electron beam and the emitted secondary and backscattered electrons used for imaging.

FourierTransformInfraredSpectroscopy
type: Class
subclass of: Spectroscopy
comment: A technique used to obtain an infrared spectrum of absorption or emission of a solid, liquid, or gas
alternative label: FTIR
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q901559
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A technique used to obtain an infrared spectrum of absorption or emission of a solid, liquid, or gas
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Fourier-transform_infrared_spectroscopy

Fractography
type: Class
subclass of: OpticalTesting
comment: Fractography is the study of fracture surfaces in order to determine the relation between the microstructure and the mechanism(s) of crack initiation and propagation and, eventually, the root cause of the fracture. Fractography qualitatively interprets the mechanisms of fracture that occur in a sample by microscopic examination of fracture surface morpholog.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Fractography is the study of fracture surfaces in order to determine the relation between the microstructure and the mechanism(s) of crack initiation and propagation and, eventually, the root cause of the fracture. Fractography qualitatively interprets the mechanisms of fracture that occur in a sample by microscopic examination of fracture surface morpholog.

FreezingPointDepressionOsmometry
type: Class
subclass of: Osmometry
comment: The general principle of freezing point depression osmometry involves the relationship between the number of moles of dissolved solute in a solution and the change in freezing point.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The general principle of freezing point depression osmometry involves the relationship between the number of moles of dissolved solute in a solution and the change in freezing point.

GalvanostaticIntermittentTitrationTechnique
type: Class
subclass of: Chronopotentiometry
comment: Electrochemical method that applies current pulses to an electrochemical cell at rest and measures the voltage response.
alternative label: GITT
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q120906986
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Electrochemical method that applies current pulses to an electrochemical cell at rest and measures the voltage response.

GammaSpectrometry
type: Class
subclass of: Spectrometry
comment: Gamma-ray spectroscopy is the qualitative study of the energy spectra of gamma-ray sources, such as in the nuclear industry, geochemical investigation, and astrophysics.[1] Gamma-ray spectrometry, on the other hand, is the method used to acquire a quantitative spectrum measurement. Most radioactive sources produce gamma rays, which are of various energies and intensities. When these emissions are detected and analyzed with a spectroscopy system, a gamma-ray energy spectrum can be produced. A detailed analysis of this spectrum is typically used to determine the identity and quantity of gamma emitters present in a gamma source, and is a vital tool in radiometric assay. The gamma spectrum is characteristic of the gamma-emitting nuclides contained in the source, just like in an optical spectrometer, the optical spectrum is characteristic of the material contained in a sample.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Gamma-ray spectroscopy is the qualitative study of the energy spectra of gamma-ray sources, such as in the nuclear industry, geochemical investigation, and astrophysics.[1] Gamma-ray spectrometry, on the other hand, is the method used to acquire a quantitative spectrum measurement. Most radioactive sources produce gamma rays, which are of various energies and intensities. When these emissions are detected and analyzed with a spectroscopy system, a gamma-ray energy spectrum can be produced. A detailed analysis of this spectrum is typically used to determine the identity and quantity of gamma emitters present in a gamma source, and is a vital tool in radiometric assay. The gamma spectrum is characteristic of the gamma-emitting nuclides contained in the source, just like in an optical spectrometer, the optical spectrum is characteristic of the material contained in a sample.

GasAdsorptionPorosimetry
type: Class
subclass of: Porosimetry
comment: Gas Adsorption Porosimetry is a method used for analyzing the surface area and porosity of materials. In this method, a gas, typically nitrogen or argon, is adsorbed onto the surface of the material at various pressures and temperatures.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Gas Adsorption Porosimetry is a method used for analyzing the surface area and porosity of materials. In this method, a gas, typically nitrogen or argon, is adsorbed onto the surface of the material at various pressures and temperatures.

Grinding
type: Class
subclass of: SamplePreparation
comment: Grinding is a machining process that involves the use of a disc-shaped grinding wheel to remove material from a workpiece. There are several types of grinding wheels, some of which include grindstones, angle grinders, die grinders and specialized grinding machines.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Grinding is a machining process that involves the use of a disc-shaped grinding wheel to remove material from a workpiece. There are several types of grinding wheels, some of which include grindstones, angle grinders, die grinders and specialized grinding machines.

HPPC
type: Class
subclass of: Chronopotentiometry
comment: Electrochemical method that measures the voltage drop of a cell resulting from a square wave current load.
alternative label: HybridPulsePowerCharacterisation
alternative label: HybridPulsePowerCharacterization
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Electrochemical method that measures the voltage drop of a cell resulting from a square wave current load.

HardnessTesting
type: Class
subclass of: MechanicalTesting
comment: A test to determine the resistance a material exhibits to permanent deformation by penetration of another harder material.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A test to determine the resistance a material exhibits to permanent deformation by penetration of another harder material.

Hazard
type: Class
comment: Set of inherent properties of a substance, mixture of substances, or a process involving substances that, under production, usage, or disposal conditions, make it capable of causing adverse effects to organisms or the environment, depending on the degree of exposure; in other words, it is a source of danger.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Set of inherent properties of a substance, mixture of substances, or a process involving substances that, under production, usage, or disposal conditions, make it capable of causing adverse effects to organisms or the environment, depending on the degree of exposure; in other words, it is a source of danger.

Holder
type: Class
subclass of: CharacterisationHardware
comment: An object which supports the specimen in the correct position for the characterisation process.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: An object which supports the specimen in the correct position for the characterisation process.

HydrodynamicVoltammetry
type: Class
subclass of: Voltammetry
comment: Voltammetry with forced flow of the solution towards the electrode surface. A linear potential scan, at sufficiently slow scan rates so as to ensure a steady state response, is usually applied. Mass transport of a redox species enhanced by convection in this way results in a greater electric current. Convective mass transfer occurs up to the diffusion-limiting layer, within which the mass transfer is controlled by diffusion. Electroactive substance depletion outside the diffusion layer is annulled by convective mass transfer, which results in steady- state sigmoidal wave-shaped current-potential curves. The forced flow can be accomplished by movement either of the solution (solution stirring, or channel flow), or of the electrode (electrode rotation or vibration).
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q17028237
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Voltammetry with forced flow of the solution towards the electrode surface. A linear potential scan, at sufficiently slow scan rates so as to ensure a steady state response, is usually applied. Mass transport of a redox species enhanced by convection in this way results in a greater electric current. Convective mass transfer occurs up to the diffusion-limiting layer, within which the mass transfer is controlled by diffusion. Electroactive substance depletion outside the diffusion layer is annulled by convective mass transfer, which results in steady- state sigmoidal wave-shaped current-potential curves. The forced flow can be accomplished by movement either of the solution (solution stirring, or channel flow), or of the electrode (electrode rotation or vibration).
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Hydrodynamic_voltammetry
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

ICI
type: Class
subclass of: Chronopotentiometry
comment: Electrochemical method that measures the voltage response of an electrochemical cell under galvanostatic conditions to short interruptions in the current.
alternative label: IntermittentCurrentInterruptionMethod
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Electrochemical method that measures the voltage response of an electrochemical cell under galvanostatic conditions to short interruptions in the current.

Impedimetry
type: Class
subclass of: ElectrochemicalTesting
comment: Measurement principle in which the complex electric impedance of a system is measured, usually as a function of a small amplitude sinusoidal electrode potential.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Measurement principle in which the complex electric impedance of a system is measured, usually as a function of a small amplitude sinusoidal electrode potential.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

InteractionVolume
type: Class
comment: The volume of material, and the surrounding environment, that interacts with the probe and generate a detectable (measurable) signal (information). In Scanning Electron Microscopy (SEM), the interaction volume is the volume of material that interacts directly with the incident electron beam, is usually much smaller than the entire specimen’s volume, and can be computed by using proper models. The interaction between the scanning probe and the sample generates a series of detectable signals (back scattered electrons, secondary electrons, x-rays, specimen current, etc.) which contain information on sample morphology, microstructure, composition, etc. In x-ray diffraction, the interaction volume is the volume of material that interacts directly with the x-ray beam and is usually smaller than the volume of the entire specimen. Depending on sample’s structure and microstructure, the interaction between the sample and the x-ray incident beam generates a secondary (reflected) beam that is measured by a detector and contains information on certain sample’s properties (e.g., crystallographic structure, phase composition, grain size, residual stress...). In some cases, (like tribological characterisations) the “sample” can also be the “probe”. When analysing a system of samples that interact each other, finding a clear definition can become a complex problem. It is important to note that, in some cases, the volume of interaction could be different from the volume of detectable signal emission. Example: in Scanning Electron Microscopy (SEM), the volume of interaction between the electron probe and the material is different from the volumes that generate the captured signal.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The volume of material, and the surrounding environment, that interacts with the probe and generate a detectable (measurable) signal (information).
example: In Scanning Electron Microscopy (SEM), the interaction volume is the volume of material that interacts directly with the incident electron beam, is usually much smaller than the entire specimen’s volume, and can be computed by using proper models. The interaction between the scanning probe and the sample generates a series of detectable signals (back scattered electrons, secondary electrons, x-rays, specimen current, etc.) which contain information on sample morphology, microstructure, composition, etc. In x-ray diffraction, the interaction volume is the volume of material that interacts directly with the x-ray beam and is usually smaller than the volume of the entire specimen. Depending on sample’s structure and microstructure, the interaction between the sample and the x-ray incident beam generates a secondary (reflected) beam that is measured by a detector and contains information on certain sample’s properties (e.g., crystallographic structure, phase composition, grain size, residual stress...).
EMMO c7b62dd7 063a 4c2a 8504 42f7264ba83f: In some cases, (like tribological characterisations) the “sample” can also be the “probe”. When analysing a system of samples that interact each other, finding a clear definition can become a complex problem. It is important to note that, in some cases, the volume of interaction could be different from the volume of detectable signal emission. Example: in Scanning Electron Microscopy (SEM), the volume of interaction between the electron probe and the material is different from the volumes that generate the captured signal.

IntermediateSample
type: Class
subclass of: Sample

IonChromatography
type: Class
subclass of: Chromatography
comment: Ion chromatography (or ion-exchange chromatography) is a form of chromatography that separates ions and ionizable polar molecules based on their affinity to the ion exchanger.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Ion chromatography (or ion-exchange chromatography) is a form of chromatography that separates ions and ionizable polar molecules based on their affinity to the ion exchanger.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Ion_chromatography

IonMobilitySpectrometry
type: Class
subclass of: Spectrometry
comment: Ion mobility spectrometry (IMS) It is a method of conducting analytical research that separates and identifies ionized molecules present in the gas phase based on the mobility of the molecules in a carrier buffer gas. Even though it is used extensively for military or security objectives, such as detecting drugs and explosives, the technology also has many applications in laboratory analysis, including studying small and big biomolecules. IMS instruments are extremely sensitive stand-alone devices, but are often coupled with mass spectrometry, gas chromatography or high-performance liquid chromatography in order to achieve a multi-dimensional separation. They come in various sizes, ranging from a few millimeters to several meters depending on the specific application, and are capable of operating under a broad range of conditions. IMS instruments such as microscale high-field asymmetric-waveform ion mobility spectrometry can be palm-portable for use in a range of applications including volatile organic compound (VOC) monitoring, biological sample analysis, medical diagnosis and food quality monitoring.
alternative label: IMS
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Ion mobility spectrometry (IMS) It is a method of conducting analytical research that separates and identifies ionized molecules present in the gas phase based on the mobility of the molecules in a carrier buffer gas. Even though it is used extensively for military or security objectives, such as detecting drugs and explosives, the technology also has many applications in laboratory analysis, including studying small and big biomolecules. IMS instruments are extremely sensitive stand-alone devices, but are often coupled with mass spectrometry, gas chromatography or high-performance liquid chromatography in order to achieve a multi-dimensional separation. They come in various sizes, ranging from a few millimeters to several meters depending on the specific application, and are capable of operating under a broad range of conditions. IMS instruments such as microscale high-field asymmetric-waveform ion mobility spectrometry can be palm-portable for use in a range of applications including volatile organic compound (VOC) monitoring, biological sample analysis, medical diagnosis and food quality monitoring.

IsothermalMicrocalorimetry
type: Class
subclass of: ThermochemicalTesting
comment: Isothermal microcalorimetry (IMC) is a laboratory method for real-time monitoring and dynamic analysis of chemical, physical and biological processes. Over a period of hours or days, IMC determines the onset, rate, extent and energetics of such processes for specimens in small ampoules (e.g. 3–20 ml) at a constant set temperature (c. 15 °C–150 °C). IMC accomplishes this dynamic analysis by measuring and recording vs. elapsed time the net rate of heat flow (μJ/s = μW) to or from the specimen ampoule, and the cumulative amount of heat (J) consumed or produced.
alternative label: IMC
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Isothermal microcalorimetry (IMC) is a laboratory method for real-time monitoring and dynamic analysis of chemical, physical and biological processes. Over a period of hours or days, IMC determines the onset, rate, extent and energetics of such processes for specimens in small ampoules (e.g. 3–20 ml) at a constant set temperature (c. 15 °C–150 °C). IMC accomplishes this dynamic analysis by measuring and recording vs. elapsed time the net rate of heat flow (μJ/s = μW) to or from the specimen ampoule, and the cumulative amount of heat (J) consumed or produced.

Laboratory
type: Class
comment: The laboratory where the whole characterisation process or some of its stages take place.
alternative label: Lab
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The laboratory where the whole characterisation process or some of its stages take place.

LevelOfAutomation
type: Class
comment: Describes the level of automation of the test.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Describes the level of automation of the test.

LevelOfExpertise
type: Class
comment: Describes the level of expertise required to carry out a process (the entire test or the data processing).
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Describes the level of expertise required to carry out a process (the entire test or the data processing).

LightScattering
type: Class
subclass of: OpticalTesting
comment: Light scattering is the way light behaves when it interacts with a medium that contains particles or the boundary between different mediums where defects or structures are present. It is different than the effects of refraction, where light undergoes a change in index of refraction as it passes from one medium to another, or reflection, where light reflects back into the same medium, both of which are governed by Snell’s law. Light scattering can be caused by factors such as the nature, texture, or specific structures of a surface and the presence of gas, liquid, or solid particles through which light propagates, as well as the nature of the light itself, of its wavelengths and polarization states. It usually results in diffuse light and can also affect the dispersion of color.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Light scattering is the way light behaves when it interacts with a medium that contains particles or the boundary between different mediums where defects or structures are present. It is different than the effects of refraction, where light undergoes a change in index of refraction as it passes from one medium to another, or reflection, where light reflects back into the same medium, both of which are governed by Snell’s law. Light scattering can be caused by factors such as the nature, texture, or specific structures of a surface and the presence of gas, liquid, or solid particles through which light propagates, as well as the nature of the light itself, of its wavelengths and polarization states. It usually results in diffuse light and can also affect the dispersion of color.

LinearChronopotentiometry
type: Class
subclass of: Chronopotentiometry
comment: Chronopotentiometry where the applied current is changed linearly.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Chronopotentiometry where the applied current is changed linearly.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: chronopotentiometry where the applied current is changed linearly

Axiom
annotated source: LinearChronopotentiometry
annotated target: chronopotentiometry where the applied current is changed linearly
source: J. M. Pingarrón et al., Terminology of electrochemical methods of analysis (IUPAC Recommendations 2019), Pure and Applied Chemistry, 4, 92, 2020, 641-694. https://doi.org/10.1515/pac-2018-0109.

LinearScanVoltammetry
type: Class
subclass of: Voltammetry
comment: Voltammetry in which the current is recorded as the electrode potential is varied linearly with time. LSV corresponds to the first half cycle of cyclic voltammetry. The peak current is expressed by the Randles-Ševčík equation. The scan is usually started at a potential where no electrode reaction occurs.
alternative label: LSV
alternative label: LinearPolarization
alternative label: LinearSweepVoltammetry
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q620700
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Voltammetry in which the current is recorded as the electrode potential is varied linearly with time. LSV corresponds to the first half cycle of cyclic voltammetry. The peak current is expressed by the Randles-Ševčík equation. The scan is usually started at a potential where no electrode reaction occurs.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Linear_sweep_voltammetry
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

MassSpectrometry
type: Class
subclass of: Spectrometry
comment: Mass spectrometry is a powerful analytical technique used to quantify known materials, to identify unknown compounds within a sample, and to elucidate the structure and chemical properties of different molecules.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Mass spectrometry is a powerful analytical technique used to quantify known materials, to identify unknown compounds within a sample, and to elucidate the structure and chemical properties of different molecules.

MeasurementParameter
type: Class
comment: Describes the main input parameters that are needed to acquire the signal.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Describes the main input parameters that are needed to acquire the signal.

MeasurementSystemAdjustment
type: Class
subclass of: CharacterisationProcedure
subclass of: Restriction
  some values from: MeasurementParameter
comment: Activity which has the goal of adjusting/tuning a measing instrument, without performing a measurement on a reference sample (which is a calibration). The output of this process can be a specific measurement parameter to be used in the characteriasation measurement process. From the International Vocabulary of Metrology (VIM): Set of operations carried out on a measuring system so that it provides prescribed indications corresponding to given values of a quantity being measured. NOTE 1: If there is any doubt that the context in which the term is being used is that of metrology, the long form “adjustment of a measuring system” might be used. NOTE 2: Types of adjustment of a measuring system include zero adjustment, offset adjustment, and span adjustment (sometimes called “gain adjustment”). NOTE 3: Adjustment of a measuring system should not be confused with calibration, which is sometimes a prerequisite for adjustment. NOTE 4: After an adjustment of a measuring system, the measuring system must usually be recalibrated.
alternative label: MeasurementParameterAdjustment
EMMO 70fe84ff 99b6 4206 a9fc 9a8931836d84: From the International Vocabulary of Metrology (VIM): Set of operations carried out on a measuring system so that it provides prescribed indications corresponding to given values of a quantity being measured. NOTE 1: If there is any doubt that the context in which the term is being used is that of metrology, the long form “adjustment of a measuring system” might be used. NOTE 2: Types of adjustment of a measuring system include zero adjustment, offset adjustment, and span adjustment (sometimes called “gain adjustment”). NOTE 3: Adjustment of a measuring system should not be confused with calibration, which is sometimes a prerequisite for adjustment. NOTE 4: After an adjustment of a measuring system, the measuring system must usually be recalibrated.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Activity which has the goal of adjusting/tuning a measing instrument, without performing a measurement on a reference sample (which is a calibration). The output of this process can be a specific measurement parameter to be used in the characteriasation measurement process.
EMMO bb49844b 45d7 4f0d 8cae 8e552cbc20d6: Adjustment

MeasurementTime
type: Class
comment: The overall time needed to acquire the measurement data.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The overall time needed to acquire the measurement data.

MechanicalTesting
type: Class
subclass of: CharacterisationTechnique
comment: Mechanical testing covers a wide range of tests, which can be divided broadly into two types: 1. those that aim to determine a material's mechanical properties, independent of geometry; 2. those that determine the response of a structure to a given action, e.g. testing of composite beams, aircraft structures to destruction, etc.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Mechanical testing covers a wide range of tests, which can be divided broadly into two types: 1. those that aim to determine a material's mechanical properties, independent of geometry; 2. those that determine the response of a structure to a given action, e.g. testing of composite beams, aircraft structures to destruction, etc.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Mechanical_testing

MembraneOsmometry
type: Class
subclass of: Osmometry
comment: In the membrane osmometry technique, a pure solvent and polymer solution are separated by a semipermeable membrane, due to the higher chemical potential of the solvent in the pure solvent than in polymer solution, the solvent starts moving towards the polymer solution.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: In the membrane osmometry technique, a pure solvent and polymer solution are separated by a semipermeable membrane, due to the higher chemical potential of the solvent in the pure solvent than in polymer solution, the solvent starts moving towards the polymer solution.

MercuryPorosimetry
type: Class
subclass of: Porosimetry
comment: A method used to measure the pore size distribution and total pore volume of solid materials by infiltrating mercury into the pores under controlled pressure conditions and analyzing the amount of mercury intrusion.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A method used to measure the pore size distribution and total pore volume of solid materials by infiltrating mercury into the pores under controlled pressure conditions and analyzing the amount of mercury intrusion.

Microscopy
type: Class
subclass of: CharacterisationTechnique
comment: Microscopy is a category of characterization techniques which probe and map the surface and sub-surface structure of a material. These techniques can use photons, electrons, ions or physical cantilever probes to gather data about a sample's structure on a range of length scales.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Microscopy is a category of characterization techniques which probe and map the surface and sub-surface structure of a material. These techniques can use photons, electrons, ions or physical cantilever probes to gather data about a sample's structure on a range of length scales.

Milling
type: Class
subclass of: SamplePreparation
comment: Milling is a machining process that involves the use of a milling machine to remove material from a workpiece. Milling machines feature cutting blades that rotate while they press against the workpiece.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Milling is a machining process that involves the use of a milling machine to remove material from a workpiece. Milling machines feature cutting blades that rotate while they press against the workpiece.

Mounting
type: Class
subclass of: SamplePreparation
subclass of: Restriction
  some values from: Holder
comment: The sample is mounted on a holder.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The sample is mounted on a holder.

Nanoindentation
type: Class
subclass of: MechanicalTesting
comment: Nanoindentation (known also as nanoindentation test) is a method for testing the hardness and related mechanical properties of materials, facilitated by high-precision instrumentation in the nanometer scale, as well as analytical and computational algorithms for result evaluation. By definition, when someone performs nanoindentation, it refers to either quasistatic or continuous stiffness measurement. However, in reality with a nanoindenter it is also possible to perform scratch testing, scanning probe microscopy, and apply non-contact surface energy mapping, which can also be called nanoindentation, because they are measurements conducted using an nanoindenter.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Nanoindentation (known also as nanoindentation test) is a method for testing the hardness and related mechanical properties of materials, facilitated by high-precision instrumentation in the nanometer scale, as well as analytical and computational algorithms for result evaluation.
example: By definition, when someone performs nanoindentation, it refers to either quasistatic or continuous stiffness measurement. However, in reality with a nanoindenter it is also possible to perform scratch testing, scanning probe microscopy, and apply non-contact surface energy mapping, which can also be called nanoindentation, because they are measurements conducted using an nanoindenter.

NeutronSpinEchoSpectroscopy
type: Class
subclass of: Spectroscopy
comment: Neutron spin echo spectroscopy is a high resolution inelastic neutron scattering method probing nanosecond dynamics. Neutron spin echo (NSE) spectroscopy uses the precession of neutron spins in a magnetic field to measure the energy transfer at the sample and decouples the energy resolution from beam characteristics like monochromatisation and collimation.
alternative label: NSE
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Neutron spin echo spectroscopy is a high resolution inelastic neutron scattering method probing nanosecond dynamics. Neutron spin echo (NSE) spectroscopy uses the precession of neutron spins in a magnetic field to measure the energy transfer at the sample and decouples the energy resolution from beam characteristics like monochromatisation and collimation.

Nexafs
type: Class
subclass of: Spectroscopy
comment: Near edge X-ray absorption fine structure (NEXAFS), also known as X-ray absorption near edge structure (XANES), is a type of absorption spectroscopy that indicates the features in the X-ray absorption spectra (XAS) of condensed matter due to the photoabsorption cross section for electronic transitions from an atomic core level to final states in the energy region of 50–100 eV above the selected atomic core level ionization energy, where the wavelength of the photoelectron is larger than the interatomic distance between the absorbing atom and its first neighbour atoms.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Near edge X-ray absorption fine structure (NEXAFS), also known as X-ray absorption near edge structure (XANES), is a type of absorption spectroscopy that indicates the features in the X-ray absorption spectra (XAS) of condensed matter due to the photoabsorption cross section for electronic transitions from an atomic core level to final states in the energy region of 50–100 eV above the selected atomic core level ionization energy, where the wavelength of the photoelectron is larger than the interatomic distance between the absorbing atom and its first neighbour atoms.

NormalPulseVoltammetry
type: Class
subclass of: Voltammetry
comment: Voltammetry in which potential pulses of amplitude increasing by a constant increment and with a pulse width of 2 to 200 ms are superimposed on a constant initial potential. Normal pulse polarography is NPV in which a dropping mercury electrode is used as the working electrode. A pulse is applied just before the mechanically enforced end of the drop. The pulse width is usually 10 to 20 % of the drop time. The drop dislodgment is synchro- nized with current sampling, which is carried out just before the end of the pulse, as in NPV. Sigmoidal wave-shaped voltammograms are obtained. The current is sampled at the end of the pulse and then plotted versus the potential of the pulse. The current is sampled just before the end of the pulse, when the charging current is greatly diminished. In this way, the ratio of faradaic current to charging current is enhanced and the negative influence of charging current is partially eliminated. Due to the improved signal (faradaic current) to noise (charging current) ratio, the limit of detec- tion is lowered. The sensitivity of NPV is not affected by the reversibility of the electrode reaction of the analyte.
alternative label: NPV
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Voltammetry in which potential pulses of amplitude increasing by a constant increment and with a pulse width of 2 to 200 ms are superimposed on a constant initial potential. Normal pulse polarography is NPV in which a dropping mercury electrode is used as the working electrode. A pulse is applied just before the mechanically enforced end of the drop. The pulse width is usually 10 to 20 % of the drop time. The drop dislodgment is synchro- nized with current sampling, which is carried out just before the end of the pulse, as in NPV. Sigmoidal wave-shaped voltammograms are obtained. The current is sampled at the end of the pulse and then plotted versus the potential of the pulse. The current is sampled just before the end of the pulse, when the charging current is greatly diminished. In this way, the ratio of faradaic current to charging current is enhanced and the negative influence of charging current is partially eliminated. Due to the improved signal (faradaic current) to noise (charging current) ratio, the limit of detec- tion is lowered. The sensitivity of NPV is not affected by the reversibility of the electrode reaction of the analyte.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

NuclearMagneticResonance
type: Class
subclass of: Spectroscopy
comment: Nuclear magnetic resonance spectroscopy, most commonly known as NMR spectroscopy or magnetic resonance spectroscopy (MRS), is a spectroscopic technique to observe local magnetic fields around atomic nuclei. This spectroscopy is based on the measurement of absorption of electromagnetic radiations in the radio frequency region from roughly 4 to 900 MHz. Absorption of radio waves in the presence of magnetic field is accompanied by a special type of nuclear transition, and for this reason, such type of spectroscopy is known as Nuclear Magnetic Resonance Spectroscopy. The sample is placed in a magnetic field and the NMR signal is produced by excitation of the nuclei sample with radio waves into nuclear magnetic resonance, which is detected with sensitive radio receivers. The intramolecular magnetic field around an atom in a molecule changes the resonance frequency, thus giving access to details of the electronic structure of a molecule and its individual functional groups. As the fields are unique or highly characteristic to individual compounds, in modern organic chemistry practice, NMR spectroscopy is the definitive method to identify monomolecular organic compounds.
alternative label: Magnetic resonance spectroscopy (MRS)
alternative label: NMR
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Nuclear magnetic resonance spectroscopy, most commonly known as NMR spectroscopy or magnetic resonance spectroscopy (MRS), is a spectroscopic technique to observe local magnetic fields around atomic nuclei. This spectroscopy is based on the measurement of absorption of electromagnetic radiations in the radio frequency region from roughly 4 to 900 MHz. Absorption of radio waves in the presence of magnetic field is accompanied by a special type of nuclear transition, and for this reason, such type of spectroscopy is known as Nuclear Magnetic Resonance Spectroscopy. The sample is placed in a magnetic field and the NMR signal is produced by excitation of the nuclei sample with radio waves into nuclear magnetic resonance, which is detected with sensitive radio receivers. The intramolecular magnetic field around an atom in a molecule changes the resonance frequency, thus giving access to details of the electronic structure of a molecule and its individual functional groups. As the fields are unique or highly characteristic to individual compounds, in modern organic chemistry practice, NMR spectroscopy is the definitive method to identify monomolecular organic compounds.

OpenCircuitHold
type: Class
subclass of: Potentiometry
comment: A process in which the electric current is kept constant at 0 (i.e., open-circuit conditions).
alternative label: OCVHold
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A process in which the electric current is kept constant at 0 (i.e., open-circuit conditions).

Operator
type: Class
comment: The human operator who takes care of the whole characterisation method or sub-processes/stages.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The human operator who takes care of the whole characterisation method or sub-processes/stages.

OpticalMicroscopy
type: Class
subclass of: Microscopy
comment: Optical microscopy is a technique used to closely view a sample through the magnification of a lens with visible light.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Optical microscopy is a technique used to closely view a sample through the magnification of a lens with visible light.

OpticalTesting
type: Class
subclass of: CharacterisationTechnique

Osmometry
type: Class
subclass of: CharacterisationTechnique
comment: Osmometry is an advanced analytical method for determining the osmotic concentration of solutions. The osmotic – or solute – concentration of a colloidal system is expressed in osmoles (Osm) per unit of volume (Osm/L) or weight (Osm/kg).
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Osmometry is an advanced analytical method for determining the osmotic concentration of solutions. The osmotic – or solute – concentration of a colloidal system is expressed in osmoles (Osm) per unit of volume (Osm/L) or weight (Osm/kg).

OutlierRemoval
type: Class
subclass of: DataFiltering
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Outlier removal refers to the process of identifying and eliminating anomalous data points that deviate significantly from the overall pattern of a dataset. These outliers are generally considered to be observations that are unusually distant from other values and can potentially distort the results of analyses.

PhotoluminescenceMicroscopy
type: Class
subclass of: Microscopy
comment: Photoluminescence spectroscopy is a widely used technique for characterisation of the optical and electronic properties of semiconductors and molecules.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Photoluminescence spectroscopy is a widely used technique for characterisation of the optical and electronic properties of semiconductors and molecules.

Polishing
type: Class
subclass of: SamplePreparation
comment: Polishing is a machining process to achieve a smooth surface of the Sample, which uses abrasive compounds with smal particles that are embedded in a pad or wheel.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Polishing is a machining process to achieve a smooth surface of the Sample, which uses abrasive compounds with smal particles that are embedded in a pad or wheel.

Porosimetry
type: Class
subclass of: CharacterisationTechnique

PostProcessingModel
type: Class
comment: Mathematical model used to process data. The PostProcessingModel use is mainly intended to get secondary data from primary data.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Mathematical model used to process data.
EMMO c7b62dd7 063a 4c2a 8504 42f7264ba83f: The PostProcessingModel use is mainly intended to get secondary data from primary data.

PotentiometricStrippingAnalysis
type: Class
subclass of: Voltammetry
comment: Two-step electrochemical measurement in which 1) material is accumulated at an electrode and 2) the material is removed by chemical reaction or electrochemically at constant current with measurement of electrode potential. Historically for the analysis of metal ions, mercury ions were added to the test solution to form a mercury amalgam when reduced. Alternatively, an HMDE or MFE was used and the oxidizing agent added after amalgam formation. However, the toxicity of mercury and its compounds have all but precluded the present-day use of mercury. The accumulation is similar to that used in stripping voltammetry. The stripping potentiogram shows staircase curves of potential as a function of time. Frequently, the first derivative is displayed (dE/dt=f(t)), as this produces peak-shaped signals. The time between transitions (peaks) is proportional to the concentration of analyte in the test solution. The time between changes in potential in step 2 is related to the concentration of analyte in the solution.
alternative label: PSA
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Two-step electrochemical measurement in which 1) material is accumulated at an electrode and 2) the material is removed by chemical reaction or electrochemically at constant current with measurement of electrode potential. Historically for the analysis of metal ions, mercury ions were added to the test solution to form a mercury amalgam when reduced. Alternatively, an HMDE or MFE was used and the oxidizing agent added after amalgam formation. However, the toxicity of mercury and its compounds have all but precluded the present-day use of mercury. The accumulation is similar to that used in stripping voltammetry. The stripping potentiogram shows staircase curves of potential as a function of time. Frequently, the first derivative is displayed (dE/dt=f(t)), as this produces peak-shaped signals. The time between transitions (peaks) is proportional to the concentration of analyte in the test solution. The time between changes in potential in step 2 is related to the concentration of analyte in the solution.

Axiom
annotated source: PotentiometricStrippingAnalysis
annotated target: Two-step electrochemical measurement in which 1) material is accumulated at an electrode and 2) the material is removed by chemical reaction or electrochemically at constant current with measurement of electrode potential. Historically for the analysis of metal ions, mercury ions were added to the test solution to form a mercury amalgam when reduced. Alternatively, an HMDE or MFE was used and the oxidizing agent added after amalgam formation. However, the toxicity of mercury and its compounds have all but precluded the present-day use of mercury. The accumulation is similar to that used in stripping voltammetry. The stripping potentiogram shows staircase curves of potential as a function of time. Frequently, the first derivative is displayed (dE/dt=f(t)), as this produces peak-shaped signals. The time between transitions (peaks) is proportional to the concentration of analyte in the test solution. The time between changes in potential in step 2 is related to the concentration of analyte in the solution.
source: J. M. Pingarrón et al., Terminology of electrochemical methods of analysis (IUPAC Recommendations 2019), Pure and Applied Chemistry, 4, 92, 2020, 641-694. https://doi.org/10.1515/pac-2018-0109

Potentiometry
type: Class
subclass of: ElectrochemicalTesting
comment: Method of electroanalytical chemistry based on measurement of an electrode potential. Potentiometric methods are used to measure the electrochemical potentials of a metallic structure in a given environment. For measurements using ion-selective electrodes, the measurement is made under equilibrium conditions what means that the macroscopic electric current is zero and the concentrations of all species are uniform throughout the solution. The indicator electrode is in direct contact with the analyte solution, whereas the reference electrode is usually separated from the analyte solution by a salt bridge. The potential difference between the indicator and reference electrodes is normally directly proportional to the logarithm of the activity (concentration) of the analyte in the solution (Nernst equation). See also ion selective electrode.
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q900632
EMMO 50c298c2 55a2 4068 b3ac 4e948c33181f: https://www.electropedia.org/iev/iev.nsf/display
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Method of electroanalytical chemistry based on measurement of an electrode potential. Potentiometric methods are used to measure the electrochemical potentials of a metallic structure in a given environment. For measurements using ion-selective electrodes, the measurement is made under equilibrium conditions what means that the macroscopic electric current is zero and the concentrations of all species are uniform throughout the solution. The indicator electrode is in direct contact with the analyte solution, whereas the reference electrode is usually separated from the analyte solution by a salt bridge. The potential difference between the indicator and reference electrodes is normally directly proportional to the logarithm of the activity (concentration) of the analyte in the solution (Nernst equation). See also ion selective electrode.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

PreparedSample
type: Class
subclass of: Sample
disjoint with: ReferenceSample
comment: The sample after a preparation process.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The sample after a preparation process.

PrimaryData
type: Class
subclass of: CharacterisationData
comment: Data resulting of a pre-processing of raw data, applying corrections to normalize/harmonize, in order to prepare them for the post-processing. Examples include: baseline subtraction, noise reduction , X and Y axes correction.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Data resulting of a pre-processing of raw data, applying corrections to normalize/harmonize, in order to prepare them for the post-processing.
example: Baseline subtraction, noise reduction , X and Y axes correction.

Probe
type: Class
subclass of: CharacterisationHardware
comment: Probe is the physical tool (i.e., a disturbance, primary solicitation, or a gadget), controlled over time, that generates measurable fields that interact with the sample to acquire information on the specimen’s behaviour and properties. In dynamic light scattering, temporal fluctuations of backscattered light due to Brownian motion and flow of nanoparticles are the probe, resolved as function of pathlength in the sample. From fluctuation analysis (intensity correlations) and the wavelength of light in the medium, the (distribution of) diffusion coefficient(s) can be measured during flow. The Stokes-Einstein relation yields the particle size characteristics. In electron microscopy (SEM or TEM), the probe is a beam of electrons with known energy that is focused (and scanned) on the sample’s surface with a well-defined beam-size and scanning algorithm. In mechanical testing, the probe is a the tip plus a force actuator, which is designed to apply a force over-time on a sample. Many variants can be defined depending on way the force is applied (tensile/compressive uniaxial tests, bending test, indentation test) and its variation with time (static tests, dynamic/cyclic tests, impact tests, etc…) In spectroscopic methods, the probe is a beam of light with pre-defined energy (for example in the case of laser beam for Raman measurements) or pre-defined polarization (for example in the case of light beam for Spectroscopic Ellipsometry methods), that will be properly focused on the sample’s surface with a welldefined geometry (specific angle of incidence). In x-ray diffraction, the probe is a beam of x-rays with known energy that is properly focused on the sample’s surface with a well-defined geometry.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Probe is the physical tool (i.e., a disturbance, primary solicitation, or a gadget), controlled over time, that generates measurable fields that interact with the sample to acquire information on the specimen’s behaviour and properties.
example: In dynamic light scattering, temporal fluctuations of backscattered light due to Brownian motion and flow of nanoparticles are the probe, resolved as function of pathlength in the sample. From fluctuation analysis (intensity correlations) and the wavelength of light in the medium, the (distribution of) diffusion coefficient(s) can be measured during flow. The Stokes-Einstein relation yields the particle size characteristics. In electron microscopy (SEM or TEM), the probe is a beam of electrons with known energy that is focused (and scanned) on the sample’s surface with a well-defined beam-size and scanning algorithm. In mechanical testing, the probe is a the tip plus a force actuator, which is designed to apply a force over-time on a sample. Many variants can be defined depending on way the force is applied (tensile/compressive uniaxial tests, bending test, indentation test) and its variation with time (static tests, dynamic/cyclic tests, impact tests, etc…) In spectroscopic methods, the probe is a beam of light with pre-defined energy (for example in the case of laser beam for Raman measurements) or pre-defined polarization (for example in the case of light beam for Spectroscopic Ellipsometry methods), that will be properly focused on the sample’s surface with a welldefined geometry (specific angle of incidence). In x-ray diffraction, the probe is a beam of x-rays with known energy that is properly focused on the sample’s surface with a well-defined geometry.

ProbeSampleInteraction
type: Class
subclass of: Restriction
  some values from: Probe
subclass of: Restriction
  some values from: Sample
subclass of: Restriction
  some values from: Signal
comment: Process representing the interaction between the Probe and the Sample (with a certain Interaction Volume) which generates a Signal.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Process representing the interaction between the Probe and the Sample (with a certain Interaction Volume) which generates a Signal.

Profilometry
type: Class
subclass of: CharacterisationTechnique
comment: Profilometry is a technique used to extract topographical data from a surface. This can be a single point, a line scan or even a full three dimensional scan. The purpose of profilometry is to get surface morphology, step heights and surface roughness.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Profilometry is a technique used to extract topographical data from a surface. This can be a single point, a line scan or even a full three dimensional scan. The purpose of profilometry is to get surface morphology, step heights and surface roughness.

PseudoOpenCircuitVoltageMethod
type: Class
subclass of: Chronopotentiometry
comment: A technique used to measure the voltage of a cell under a low applied current as an estimate for the open-circuit voltage.
alternative label: PseudoOCV
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A technique used to measure the voltage of a cell under a low applied current as an estimate for the open-circuit voltage.

PulsedElectroacousticMethod
type: Class
subclass of: ChargeDistribution
comment: The pulsed electroacoustic (PEA) method is an established method for space charge measurements in polymeric dielectrics.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The pulsed electroacoustic (PEA) method is an established method for space charge measurements in polymeric dielectrics.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1007/s10832-023-00332-y

RamanSpectroscopy
type: Class
subclass of: Spectroscopy
comment: Raman spectroscopy (/ˈrɑːmən/) (named after physicist C. V. Raman) is a spectroscopic technique typically used to determine vibrational modes of molecules, although rotational and other low-frequency modes of systems may also be observed. Raman spectroscopy is commonly used in chemistry to provide a structural fingerprint by which molecules can be identified. Raman spectroscopy relies upon inelastic scattering of photons, known as Raman scattering. A source of monochromatic light, usually from a laser in the visible, near infrared, or near ultraviolet range is used, although X-rays can also be used. The laser light interacts with molecular vibrations, phonons or other excitations in the system, resulting in the energy of the laser photons being shifted up or down. The shift in energy gives information about the vibrational modes in the system. Infrared spectroscopy typically yields similar yet complementary information. Typically, a sample is illuminated with a laser beam. Electromagnetic radiation from the illuminated spot is collected with a lens and sent through a monochromator. Elastic scattered radiation at the wavelength corresponding to the laser line (Rayleigh scattering) is filtered out by either a notch filter, edge pass filter, or a band pass filter, while the rest of the collected light is dispersed onto a detector.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Raman spectroscopy (/ˈrɑːmən/) (named after physicist C. V. Raman) is a spectroscopic technique typically used to determine vibrational modes of molecules, although rotational and other low-frequency modes of systems may also be observed. Raman spectroscopy is commonly used in chemistry to provide a structural fingerprint by which molecules can be identified. Raman spectroscopy relies upon inelastic scattering of photons, known as Raman scattering. A source of monochromatic light, usually from a laser in the visible, near infrared, or near ultraviolet range is used, although X-rays can also be used. The laser light interacts with molecular vibrations, phonons or other excitations in the system, resulting in the energy of the laser photons being shifted up or down. The shift in energy gives information about the vibrational modes in the system. Infrared spectroscopy typically yields similar yet complementary information. Typically, a sample is illuminated with a laser beam. Electromagnetic radiation from the illuminated spot is collected with a lens and sent through a monochromator. Elastic scattered radiation at the wavelength corresponding to the laser line (Rayleigh scattering) is filtered out by either a notch filter, edge pass filter, or a band pass filter, while the rest of the collected light is dispersed onto a detector.

Rationale
type: Class
comment: A set of reasons or a logical basis for a decision or belief.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A set of reasons or a logical basis for a decision or belief.

RawData
type: Class
subclass of: CharacterisationData
comment: Direct output of the equipment with the manufacturer’s software including automatic pre-processing that is not modified by the user once the acquisition method is defined and the equipment calibrated. The raw data is a set of (unprocessed) data that is given directly as output from the detector, usually expressed as a function of time or position, or photon energy. In mechanical testing, examples of raw data are raw-force, raw-displacement, coordinates as function of time. In spectroscopic testing, the raw data are light intensity, or refractive index, or optical absorption as a function of the energy (or wavelength) of the incident light beam. In some cases, raw data can be considered to have already some level of data processing, e.g., in electron microscopy a “raw image” that is formed on the screen is already result from multiple processing after the signal is acquired by the detector.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Direct output of the equipment with the manufacturer’s software including automatic pre-processing that is not modified by the user once the acquisition method is defined and the equipment calibrated. The raw data is a set of (unprocessed) data that is given directly as output from the detector, usually expressed as a function of time or position, or photon energy.
example: In mechanical testing, examples of raw data are raw-force, raw-displacement, coordinates as function of time. In spectroscopic testing, the raw data are light intensity, or refractive index, or optical absorption as a function of the energy (or wavelength) of the incident light beam.
EMMO c7b62dd7 063a 4c2a 8504 42f7264ba83f: In some cases, raw data can be considered to have already some level of data processing, e.g., in electron microscopy a “raw image” that is formed on the screen is already result from multiple processing after the signal is acquired by the detector.

RawSample
type: Class
subclass of: Sample

ReferenceSample
type: Class
subclass of: Sample
comment: Material, sufficiently homogeneous and stable with respect to one or more specified properties, which has been established to be fit for its intended use in a measurement process. From the International Vocabulary of Metrology (VIM): Material, sufficiently homogeneous and stable with reference to one or more specified properties, which has been established to be fit for its intended use in measurement or in examination. NOTE 1 Reference materials can be certified reference materials or reference materials without a certified property value. NOTE 2 For a reference material to be used as a measurement standard for calibration purposes it needs to be a certified reference material. NOTE 3 Reference materials can be used for measurement precision evaluation and quality control. EXAMPLE Human serum without an assigned quantity value for the amount-of-substance concentration of the inherent cholesterol, used for quality control. NOTE 4 Properties of reference materials can be quantities or nominal properties. NOTE 5 A reference material is sometimes incorporated into a specially fabricated device. EXAMPLE Spheres of uniform size mounted on a microscope slide. NOTE 6 Some reference materials have assigned values in a unit outside the SI. Such materials include vaccines to which International Units (IU) have been assigned by the World Health Organization. NOTE 7 A given reference material can only be used for one purpose in a measurement, either calibration or quality control, but not both. NOTE 8 ISO/REMCO has an analogous definition but uses the term “measurement process” (ISO Guide 30, Reference materials – Selected terms and definitions, definition 2.1.1) for both measurement and examination. From [ISO 17858:2007]: Quality control sample used to determine accuracy and precision of method.
alternative label: Certified Reference Material
alternative label: Reference material
alternative label: ReferenceSpecimen
EMMO 70fe84ff 99b6 4206 a9fc 9a8931836d84: From the International Vocabulary of Metrology (VIM): Material, sufficiently homogeneous and stable with reference to one or more specified properties, which has been established to be fit for its intended use in measurement or in examination. NOTE 1 Reference materials can be certified reference materials or reference materials without a certified property value. NOTE 2 For a reference material to be used as a measurement standard for calibration purposes it needs to be a certified reference material. NOTE 3 Reference materials can be used for measurement precision evaluation and quality control. EXAMPLE Human serum without an assigned quantity value for the amount-of-substance concentration of the inherent cholesterol, used for quality control. NOTE 4 Properties of reference materials can be quantities or nominal properties. NOTE 5 A reference material is sometimes incorporated into a specially fabricated device. EXAMPLE Spheres of uniform size mounted on a microscope slide. NOTE 6 Some reference materials have assigned values in a unit outside the SI. Such materials include vaccines to which International Units (IU) have been assigned by the World Health Organization. NOTE 7 A given reference material can only be used for one purpose in a measurement, either calibration or quality control, but not both. NOTE 8 ISO/REMCO has an analogous definition but uses the term “measurement process” (ISO Guide 30, Reference materials – Selected terms and definitions, definition 2.1.1) for both measurement and examination. From [ISO 17858:2007]: Quality control sample used to determine accuracy and precision of method.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Material, sufficiently homogeneous and stable with respect to one or more specified properties, which has been established to be fit for its intended use in a measurement process.
EMMO bb49844b 45d7 4f0d 8cae 8e552cbc20d6: Reference material

Sample
type: Class
comment: Portion of material selected from a larger quantity of material. The term needs to be qualified, e.g., bulk sample, representative sample, primary sample, bulked sample, test sample, etc. The term 'sample' implies the existence of a sampling error, i.e., the results obtained on the portions taken are only estimates of the concentration of a constituent or the quantity of a property present in the parent material. Sample and Specimen are often used interchangeably. However in some cases the term Specimen is used to specify a portion taken under conditions such that the sampling variability cannot be assessed (usually because the population is changing), and is assumed, for convenience, to be zero.
alternative label: Specimen
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Portion of material selected from a larger quantity of material. The term needs to be qualified, e.g., bulk sample, representative sample, primary sample, bulked sample, test sample, etc. The term 'sample' implies the existence of a sampling error, i.e., the results obtained on the portions taken are only estimates of the concentration of a constituent or the quantity of a property present in the parent material.
EMMO c7b62dd7 063a 4c2a 8504 42f7264ba83f: Sample and Specimen are often used interchangeably. However in some cases the term Specimen is used to specify a portion taken under conditions such that the sampling variability cannot be assessed (usually because the population is changing), and is assumed, for convenience, to be zero.

SampleExtraction
type: Class
subclass of: CharacterisationProcedure
subclass of: Restriction
subclass of: Restriction
  some values from: Sample
comment: Act of extracting a portion (amount) of material from a larger quantity of material. This operation results in obtaining a sample representative of the batch with respect to the property or properties being investigated. The term can be used to cover either a unit of supply or a portion for analysis. The portion taken may consist of one or more sub-samples and the batch may be the population from which the sample is taken.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Act of extracting a portion (amount) of material from a larger quantity of material. This operation results in obtaining a sample representative of the batch with respect to the property or properties being investigated.
EMMO c7b62dd7 063a 4c2a 8504 42f7264ba83f: The term can be used to cover either a unit of supply or a portion for analysis. The portion taken may consist of one or more sub-samples and the batch may be the population from which the sample is taken.

SampleExtractionByCutting
type: Class
subclass of: SampleExtraction

SampleExtractionInstrument
type: Class
subclass of: CharacterisationHardware

SampleInspection
type: Class
subclass of: CharacterisationProcedure
subclass of: Restriction
  some values from: Sample
subclass of: Restriction
  some values from: SampleInspectionInstrument
subclass of: Restriction
  some values from: SampleInspectionParameter
subclass of: Restriction
  some values from: CharacterisationData
comment: Analysis of the sample in order to determine information that are relevant for the characterisation method. In the Nanoindentation method, the Scanning Electron Microscope to determine the indentation area is a SampleInspection.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Analysis of the sample in order to determine information that are relevant for the characterisation method.
example: In the Nanoindentation method, the Scanning Electron Microscope to determine the indentation area is a SampleInspection.

SampleInspectionInstrument
type: Class
subclass of: CharacterisationHardware

SampleInspectionParameter
type: Class
comment: Parameter used for the sample inspection process.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Parameter used for the sample inspection process.

SamplePreparation
type: Class
subclass of: CharacterisationProcedure
subclass of: Restriction
  some values from: SamplePreparationInstrument
subclass of: Restriction
  some values from: Sample
subclass of: Restriction
  some values from: SamplePreparationParameter
subclass of: Restriction
  some values from: Sample
comment: Sample preparation processes (e.g., machining, polishing, cutting to size, etc.) before actual observation and measurement.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Sample preparation processes (e.g., machining, polishing, cutting to size, etc.) before actual observation and measurement.

SamplePreparationByCutting
type: Class
subclass of: SamplePreparation

SamplePreparationInstrument
type: Class
subclass of: CharacterisationHardware

SamplePreparationParameter
type: Class
comment: Parameter used for the sample preparation process.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Parameter used for the sample preparation process.

SampledDCPolarography
type: Class
subclass of: DCPolarography
comment: DC polarography with current sampling at the end of each drop life mechanically enforced by a knocker at a preset drop time value. The current sampling and mechanical drop dislodge are synchronized. In this way, the ratio of faradaic current to double layer charging current is enhanced and the negative influence of charging current is partially eliminated. Due to the improved signal (faradaic current) to noise (charging current) ratio, the limit of detection is lowered.
alternative label: TASTPolarography
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: DC polarography with current sampling at the end of each drop life mechanically enforced by a knocker at a preset drop time value. The current sampling and mechanical drop dislodge are synchronized. In this way, the ratio of faradaic current to double layer charging current is enhanced and the negative influence of charging current is partially eliminated. Due to the improved signal (faradaic current) to noise (charging current) ratio, the limit of detection is lowered.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

ScanningAugerElectronMicroscopy
type: Class
subclass of: Microscopy
comment: Auger electron spectroscopy (AES or simply Auger) is a surface analysis technique that uses an electron beam to excite electrons on atoms in the particle. Atoms that are excited by the electron beam can emit “Auger” electrons. AES measures the kinetic energies of the emitted electrons. The energy of the emitted electrons is characteristic of elements present at the surface and near the surface of a sample.
alternative label: AES
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Auger electron spectroscopy (AES or simply Auger) is a surface analysis technique that uses an electron beam to excite electrons on atoms in the particle. Atoms that are excited by the electron beam can emit “Auger” electrons. AES measures the kinetic energies of the emitted electrons. The energy of the emitted electrons is characteristic of elements present at the surface and near the surface of a sample.

ScanningElectronMicroscopy
type: Class
subclass of: Microscopy
comment: The scanning electron microscope (SEM) uses a focused beam of high-energy electrons to generate a variety of signals at the surface of solid specimens. The signals that derive from electron-sample interactions reveal information about the sample including external morphology (texture), chemical composition, and crystalline structure and orientation of materials making up the sample.
alternative label: SEM
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: The scanning electron microscope (SEM) uses a focused beam of high-energy electrons to generate a variety of signals at the surface of solid specimens. The signals that derive from electron-sample interactions reveal information about the sample including external morphology (texture), chemical composition, and crystalline structure and orientation of materials making up the sample.

ScanningKelvinProbe
type: Class
subclass of: Microscopy
comment: Scanning Kelvin probe (SKP) and scanning Kelvin probe force microscopy (SKPFM) are probe techniques which permit mapping of topography and Volta potential distribution on electrode surfaces. It measures the surface electrical potential of a sample without requiring an actual physical contact.
alternative label: SKB
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Scanning Kelvin probe (SKP) and scanning Kelvin probe force microscopy (SKPFM) are probe techniques which permit mapping of topography and Volta potential distribution on electrode surfaces. It measures the surface electrical potential of a sample without requiring an actual physical contact.

ScanningProbeMicroscopy
type: Class
subclass of: Microscopy
comment: Scanning probe microscopy (SPM) is a branch of microscopy that forms images of surfaces using a physical probe that scans the specimen.
alternative label: SPM
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Scanning probe microscopy (SPM) is a branch of microscopy that forms images of surfaces using a physical probe that scans the specimen.

ScanningTunnelingMicroscopy
type: Class
subclass of: Microscopy
comment: Scanning Tunneling Microscopy, or STM, is an imaging technique used to obtain ultra-high resolution images at the atomic scale, without using light or electron beams.
alternative label: STM
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Scanning Tunneling Microscopy, or STM, is an imaging technique used to obtain ultra-high resolution images at the atomic scale, without using light or electron beams.

ScatteringAndDiffraction
type: Class
subclass of: CharacterisationTechnique

SecondaryData
type: Class
subclass of: CharacterisationData
comment: Data resulting from the application of post-processing or model generation to other data. Examples include: deconvoluted curves, intensity maps.
alternative label: Elaborated data
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Data resulting from the application of post-processing or model generation to other data.
example: Deconvoluted curves, intensity maps.

SecondaryIonMassSpectrometry
type: Class
subclass of: Spectrometry
comment: Secondary-ion mass spectrometry (SIMS) is a technique used to analyze the composition of solid surfaces and thin films by sputtering the surface of the specimen with a focused primary ion beam and collecting and analyzing ejected secondary ions.
alternative label: SIMS
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Secondary-ion mass spectrometry (SIMS) is a technique used to analyze the composition of solid surfaces and thin films by sputtering the surface of the specimen with a focused primary ion beam and collecting and analyzing ejected secondary ions.

ShearOrTorsionTesting
type: Class
subclass of: MechanicalTesting

Signal
type: Class
subclass of: CharacterisationData
comment: Result (effect) of the interaction between the sample and the probe, which usually is a measurable and quantifiable quantity. According to UPAC Compendium of Chemical Terminology, a “signal” is “A representation of a quantity within an analytical instrument” (https://goldbook.iupac.org/terms/view/S05661). Signal is usually emitted from a characteristic “emission” volume, which can be different from the sample/probe “interaction” volume and can be usually quantified using proper physics equations and/or modelling of the interaction mechanisms.
EMMO 70fe84ff 99b6 4206 a9fc 9a8931836d84: According to UPAC Compendium of Chemical Terminology, a “signal” is “A representation of a quantity within an analytical instrument” (https://goldbook.iupac.org/terms/view/S05661).
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Result (effect) of the interaction between the sample and the probe, which usually is a measurable and quantifiable quantity.
EMMO c7b62dd7 063a 4c2a 8504 42f7264ba83f: Signal is usually emitted from a characteristic “emission” volume, which can be different from the sample/probe “interaction” volume and can be usually quantified using proper physics equations and/or modelling of the interaction mechanisms.

Spectrometry
type: Class
subclass of: CharacterisationTechnique
comment: Spectroscopic techniques are numerous and varied, but all involve measuring the response of a material to different frequencies of electromagnetic radiation. Depending on the technique used, material characterization may be based on the absorption, emission, impedance, or reflection of incident energy by a sample.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Spectroscopic techniques are numerous and varied, but all involve measuring the response of a material to different frequencies of electromagnetic radiation. Depending on the technique used, material characterization may be based on the absorption, emission, impedance, or reflection of incident energy by a sample.

Spectroscopy
type: Class
subclass of: CharacterisationTechnique
comment: Spectroscopy is a category of characterization techniques which use a range of principles to reveal the chemical composition, composition variation, crystal structure and photoelectric properties of materials.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Spectroscopy is a category of characterization techniques which use a range of principles to reveal the chemical composition, composition variation, crystal structure and photoelectric properties of materials.

SquareWaveVoltammetry
type: Class
subclass of: Voltammetry
comment: Voltammetry in which a square-wave potential waveform is superimposed on an underlying linearly varying potential ramp or staircase ramp. Most instruments show plots of the current at the end of the forward-going pulse and of the backward-going pulse vs. the potential, as well as their difference. This can give valuable information on the kinetics of the electrode reaction and the electrode process. The current is sampled just before the end of the forward- going pulse and of the backward-going pulse and the difference of the two sampled currents is plotted versus the applied potential of the potential or staircase ramp. The square-wave voltammogram is peak-shaped. The sensitivity of SWV depends on the reversibility of the electrode reaction of the analyte.
alternative label: OSWV
alternative label: OsteryoungSquareWaveVoltammetry
alternative label: SWV
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q4016323
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Voltammetry in which a square-wave potential waveform is superimposed on an underlying linearly varying potential ramp or staircase ramp. Most instruments show plots of the current at the end of the forward-going pulse and of the backward-going pulse vs. the potential, as well as their difference. This can give valuable information on the kinetics of the electrode reaction and the electrode process. The current is sampled just before the end of the forward- going pulse and of the backward-going pulse and the difference of the two sampled currents is plotted versus the applied potential of the potential or staircase ramp. The square-wave voltammogram is peak-shaped. The sensitivity of SWV depends on the reversibility of the electrode reaction of the analyte.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Squarewave_voltammetry
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

StepChronopotentiometry
type: Class
subclass of: Chronopotentiometry
comment: Chronopotentiometry where the applied current is changed in steps.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Chronopotentiometry where the applied current is changed in steps.

Axiom
annotated source: StepChronopotentiometry
annotated target: Chronopotentiometry where the applied current is changed in steps.
source: J. M. Pingarrón et al., Terminology of electrochemical methods of analysis (IUPAC Recommendations 2019), Pure and Applied Chemistry, 4, 92, 2020, 641-694. https://doi.org/10.1515/pac-2018-0109.

StrippingVoltammetry
type: Class
subclass of: Voltammetry
comment: Two-step electrochemical measurement in which 1) material is accumulated at an electrode and 2) the amount of an accumulated species is measured by voltammetry. The measured electric current in step 2 is related to the concentration of analyte in the solution by calibration. Anodic stripping voltammetry (ASV) was historically used to measure concentrations of metal ions in solution using cathodic accumulation with mercury to form an amalgam. Due to the toxicity of mercury and its compounds, inductively coupled plasma optical emission spectrometry and inductively coupled plasma mass spectrometry have frequently replaced ASV at mercury electrodes in the laboratory, often sacrificing the probing of speciation and lability in complex matrices. Mercury has now been replaced by non-toxic bismuth or anti- mony as films on a solid electrode support (such as glassy carbon) with equally good sensi- tivity and detection limits. Because the accumulation (pre-concentration) step can be prolonged, increasing the amount of material at the electrode, stripping voltammetry is able to measure very small concentrations of analyte. Often the product of the electrochemical stripping is identical to the analyte before the accumulation. Stripping voltammetry is a calibrated method to establish the relation between amount accumulated in a given time and the concentration of the analyte in solution. Types of stripping voltammetry refer to the kind of accumulation (e.g. adsorptive stripping voltammetry) or the polarity of the stripping electrochemistry (anodic, cathodic stripping voltammetry).
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Two-step electrochemical measurement in which 1) material is accumulated at an electrode and 2) the amount of an accumulated species is measured by voltammetry. The measured electric current in step 2 is related to the concentration of analyte in the solution by calibration. Anodic stripping voltammetry (ASV) was historically used to measure concentrations of metal ions in solution using cathodic accumulation with mercury to form an amalgam. Due to the toxicity of mercury and its compounds, inductively coupled plasma optical emission spectrometry and inductively coupled plasma mass spectrometry have frequently replaced ASV at mercury electrodes in the laboratory, often sacrificing the probing of speciation and lability in complex matrices. Mercury has now been replaced by non-toxic bismuth or anti- mony as films on a solid electrode support (such as glassy carbon) with equally good sensi- tivity and detection limits. Because the accumulation (pre-concentration) step can be prolonged, increasing the amount of material at the electrode, stripping voltammetry is able to measure very small concentrations of analyte. Often the product of the electrochemical stripping is identical to the analyte before the accumulation. Stripping voltammetry is a calibrated method to establish the relation between amount accumulated in a given time and the concentration of the analyte in solution. Types of stripping voltammetry refer to the kind of accumulation (e.g. adsorptive stripping voltammetry) or the polarity of the stripping electrochemistry (anodic, cathodic stripping voltammetry).
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Electrochemical_stripping_analysis
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

Synchrotron
type: Class
subclass of: ScatteringAndDiffraction

TensileTesting
type: Class
subclass of: MechanicalTesting
comment: Tensile testing, also known as tension testing, is a test in which a sample is subjected to a controlled tension until failure. Properties that are directly measured via a tensile test are ultimate tensile strength, breaking strength, maximum elongation and reduction in area. From these measurements the following properties can also be determined: Young's modulus, Poisson's ratio, yield strength, and strain-hardening characteristics. Uniaxial tensile testing is the most commonly used for obtaining the mechanical characteristics of isotropic materials. Some materials use biaxial tensile testing. The main difference between these testing machines being how load is applied on the materials.
alternative label: TensionTest
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Tensile testing, also known as tension testing, is a test in which a sample is subjected to a controlled tension until failure. Properties that are directly measured via a tensile test are ultimate tensile strength, breaking strength, maximum elongation and reduction in area. From these measurements the following properties can also be determined: Young's modulus, Poisson's ratio, yield strength, and strain-hardening characteristics. Uniaxial tensile testing is the most commonly used for obtaining the mechanical characteristics of isotropic materials. Some materials use biaxial tensile testing. The main difference between these testing machines being how load is applied on the materials.

ThermochemicalTesting
type: Class
subclass of: CharacterisationTechnique
comment: Thermomechanical analysis (TMA) is a technique used in thermal analysis, a branch of materials science which studies the properties of materials as they change with temperature.
alternative label: TMA
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Thermomechanical analysis (TMA) is a technique used in thermal analysis, a branch of materials science which studies the properties of materials as they change with temperature.

Thermogravimetry
type: Class
subclass of: ThermochemicalTesting
comment: Thermogravimetric analysis or thermal gravimetric analysis (TGA) is a method of thermal analysis in which the mass of a sample is measured over time as the temperature changes. This measurement provides information about physical phenomena, such as phase transitions, absorption, adsorption and desorption; as well as chemical phenomena including chemisorptions, thermal decomposition, and solid-gas reactions (e.g., oxidation or reduction).
alternative label: TGA
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Thermogravimetric analysis or thermal gravimetric analysis (TGA) is a method of thermal analysis in which the mass of a sample is measured over time as the temperature changes. This measurement provides information about physical phenomena, such as phase transitions, absorption, adsorption and desorption; as well as chemical phenomena including chemisorptions, thermal decomposition, and solid-gas reactions (e.g., oxidation or reduction).

ThreePointBendingTesting
type: Class
subclass of: MechanicalTesting
comment: Method of mechanical testing that provides values for the modulus of elasticity in bending, flexural stress, flexural strain, and the flexural stress–strain response of a material sample.
alternative label: ThreePointFlexuralTest
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q2300905
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Method of mechanical testing that provides values for the modulus of elasticity in bending, flexural stress, flexural strain, and the flexural stress–strain response of a material sample.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Three-point_flexural_test

Tomography
type: Class
subclass of: CharacterisationTechnique
comment: Tomography is imaging by sections or sectioning that uses any kind of penetrating wave. The method is used in radiology, archaeology, biology, atmospheric science, geophysics, oceanography, plasma physics, materials science, cosmochemistry, astrophysics, quantum information, and other areas of science. The word tomography is derived from Ancient Greek τόμος tomos, "slice, section" and γράφω graphō, "to write" or, in this context as well, "to describe." A device used in tomography is called a tomograph, while the image produced is a tomogram.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Tomography is imaging by sections or sectioning that uses any kind of penetrating wave. The method is used in radiology, archaeology, biology, atmospheric science, geophysics, oceanography, plasma physics, materials science, cosmochemistry, astrophysics, quantum information, and other areas of science. The word tomography is derived from Ancient Greek τόμος tomos, "slice, section" and γράφω graphō, "to write" or, in this context as well, "to describe." A device used in tomography is called a tomograph, while the image produced is a tomogram.

TransmissionElectronMicroscopy
type: Class
subclass of: Microscopy
comment: Transmission electron microscopy (TEM) is a microscopy technique in which a beam of electrons is transmitted through a specimen to form an image. The specimen is most often an ultrathin section less than 100 nm thick or a suspension on a grid. An image is formed from the interaction of the electrons with the sample as the beam is transmitted through the specimen. The image is then magnified and focused onto an imaging device, such as a fluorescent screen, a layer of photographic film, or a sensor such as a scintillator attached to a charge-coupled device.
alternative label: TEM
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Transmission electron microscopy (TEM) is a microscopy technique in which a beam of electrons is transmitted through a specimen to form an image. The specimen is most often an ultrathin section less than 100 nm thick or a suspension on a grid. An image is formed from the interaction of the electrons with the sample as the beam is transmitted through the specimen. The image is then magnified and focused onto an imaging device, such as a fluorescent screen, a layer of photographic film, or a sensor such as a scintillator attached to a charge-coupled device.

UltrasonicTesting
type: Class
subclass of: CharacterisationTechnique
comment: Ultrasonic testing (UT) is a family of non-destructive testing techniques based on the propagation of ultrasonic waves in the object or material tested. In most common UT applications, very short ultrasonic pulse-waves with center frequencies ranging from 0.1-15 MHz, and occasionally up to 50 MHz, are transmitted into materials to detect internal flaws or to characterize materials. A common example is ultrasonic thickness measurement, which tests the thickness of the test object, for example, to monitor pipework corrosion. Ultrasonic testing is often performed on steel and other metals and alloys, though it can also be used on concrete, wood and composites, albeit with less resolution. It is used in many industries including steel and aluminium construction, metallurgy, manufacturing, aerospace, automotive and other transportation sectors.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Ultrasonic testing (UT) is a family of non-destructive testing techniques based on the propagation of ultrasonic waves in the object or material tested. In most common UT applications, very short ultrasonic pulse-waves with center frequencies ranging from 0.1-15 MHz, and occasionally up to 50 MHz, are transmitted into materials to detect internal flaws or to characterize materials. A common example is ultrasonic thickness measurement, which tests the thickness of the test object, for example, to monitor pipework corrosion. Ultrasonic testing is often performed on steel and other metals and alloys, though it can also be used on concrete, wood and composites, albeit with less resolution. It is used in many industries including steel and aluminium construction, metallurgy, manufacturing, aerospace, automotive and other transportation sectors.

UltravioletVisibleSpectroscopy
type: Class
subclass of: Spectroscopy
comment: Ultraviolet–visible spectroscopy or UV/VIS refers to absorption spectroscopy or reflectance spectroscopy in the ultraviolet-visible spectral region. It is routinely used in material science and analytical chemistry, commonly carried out in gas, solutions or vacuum environment.
alternative label: UV-VIS
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Ultraviolet–visible spectroscopy or UV/VIS refers to absorption spectroscopy or reflectance spectroscopy in the ultraviolet-visible spectral region.
EMMO c7b62dd7 063a 4c2a 8504 42f7264ba83f: It is routinely used in material science and analytical chemistry, commonly carried out in gas, solutions or vacuum environment.

UserCase
type: Class
comment: High level description of the user case. It can include the properties of the material, the conditions of the environment and possibly mentioning which are the industrial sectors of reference.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: High level description of the user case. It can include the properties of the material, the conditions of the environment and possibly mentioning which are the industrial sectors of reference.

VaporPressureDepressionOsmometry
type: Class
subclass of: Osmometry
comment: Vapor pressure osmometry measures vapor pressure indirectly by measuring the change in temperature of a polymer solution on dilution by solvent vapor and is generally useful for polymers with Mn below 10,000–40,000 g/mol. When molecular weight is more than that limit, the quantity being measured becomes very small to detect.
alternative label: VPO
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Vapor pressure osmometry measures vapor pressure indirectly by measuring the change in temperature of a polymer solution on dilution by solvent vapor and is generally useful for polymers with Mn below 10,000–40,000 g/mol. When molecular weight is more than that limit, the quantity being measured becomes very small to detect.

Viscometry
type: Class
subclass of: CharacterisationTechnique
comment: Viscometry or viscosity method was one of the first methods used for determining the MW of polymers. In this method, the viscosity of polymer solution is measured, and the simplest method used is capillary viscometry by using the Ubbelohde U-tube viscometer. In this method, both the flow time of the polymer solution (t) and the flow time of the pure solvent (t0) are recorded. The ratio of the polymer solution flow time (t) to the flow time of pure solvent (t0) is equal to the ratio of their viscosities (η/η0) only if they have the same densities.
alternative label: Viscosity
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Viscometry or viscosity method was one of the first methods used for determining the MW of polymers. In this method, the viscosity of polymer solution is measured, and the simplest method used is capillary viscometry by using the Ubbelohde U-tube viscometer. In this method, both the flow time of the polymer solution (t) and the flow time of the pure solvent (t0) are recorded. The ratio of the polymer solution flow time (t) to the flow time of pure solvent (t0) is equal to the ratio of their viscosities (η/η0) only if they have the same densities.

Voltammetry
type: Class
subclass of: ElectrochemicalTesting
comment: Voltammetry is an analytical technique based on the measure of the current flowing through an electrode dipped in a solution containing electro-active compounds, while a potential scanning is imposed upon it. The current vs. potential (I-E) curve is called a voltammogram.
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q904093
EMMO 50c298c2 55a2 4068 b3ac 4e948c33181f: https://www.electropedia.org/iev/iev.nsf/display
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Voltammetry is an analytical technique based on the measure of the current flowing through an electrode dipped in a solution containing electro-active compounds, while a potential scanning is imposed upon it. The current vs. potential (I-E) curve is called a voltammogram.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Voltammetry
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

VoltammetryAtARotatingDiskElectrode
type: Class
subclass of: HydrodynamicVoltammetry
comment: Hydrodynamic voltammetry using a a rotating disc electrode, where the limiting current is described by the Levich equation.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: Hydrodynamic voltammetry using a a rotating disc electrode, where the limiting current is described by the Levich equation.
EMMO fe015383 afb3 44a6 ae86 043628697aa2: https://doi.org/10.1515/pac-2018-0109

WearTesting
type: Class
subclass of: MechanicalTesting
comment: A wear test measures the changes in conditions caused by friction, and the result is obtained from deformation, scratches, and indentations on the interacting surfaces. Wear is defined as the progressive removal of the material from a solid surface and manifested by a change in the geometry of the surface.
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A wear test measures the changes in conditions caused by friction, and the result is obtained from deformation, scratches, and indentations on the interacting surfaces. Wear is defined as the progressive removal of the material from a solid surface and manifested by a change in the geometry of the surface.

XpsVariableKinetic
type: Class
subclass of: Spectroscopy
comment: X-ray photoelectron spectroscopy (XPS), also known as ESCA (electron spectroscopy for chemical analysis) is a surface analysis technique which provides both elemental and chemical state information virtually without restriction on the type of material which can be analysed. It is a relatively simple technique where the sample is illuminated with X-rays which have enough energy to eject an electron from the atom. These ejected electrons are known as photoelectrons. The kinetic energy of these emitted electrons is characteristic of the element from which the photoelectron originated. The position and intensity of the peaks in an energy spectrum provide the desired chemical state and quantitative information. The surface sensitivity of XPS is determined by the distance that that photoelectron can travel through the material without losing any kinteic energy. These elastiaclly scattered photoelectrons contribute to the photoelectron peak, whilst photoelectrons that have been inelastically scattered, losing some kinetic energy before leaving the material, will contribute to the spectral background.
alternative label: Electron spectroscopy for chemical analysis (ESCA)
alternative label: X-ray photoelectron spectroscopy (XPS)
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: X-ray photoelectron spectroscopy (XPS), also known as ESCA (electron spectroscopy for chemical analysis) is a surface analysis technique which provides both elemental and chemical state information virtually without restriction on the type of material which can be analysed. It is a relatively simple technique where the sample is illuminated with X-rays which have enough energy to eject an electron from the atom. These ejected electrons are known as photoelectrons. The kinetic energy of these emitted electrons is characteristic of the element from which the photoelectron originated. The position and intensity of the peaks in an energy spectrum provide the desired chemical state and quantitative information. The surface sensitivity of XPS is determined by the distance that that photoelectron can travel through the material without losing any kinteic energy. These elastiaclly scattered photoelectrons contribute to the photoelectron peak, whilst photoelectrons that have been inelastically scattered, losing some kinetic energy before leaving the material, will contribute to the spectral background.

XrayDiffraction
type: Class
subclass of: ScatteringAndDiffraction
comment: A technique used to analyze the atomic and molecular structure of crystalline materials by observing the diffraction patterns produced when X-rays interact with the regular array of atoms in the crystal lattice.
alternative label: XRD
EMMO 26bf1bef d192 4da6 b0eb d2209698fb54: https://www.wikidata.org/wiki/Q12101244
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A technique used to analyze the atomic and molecular structure of crystalline materials by observing the diffraction patterns produced when X-rays interact with the regular array of atoms in the crystal lattice.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/X-ray_crystallography

XrayPowderDiffraction
type: Class
subclass of: XrayDiffraction
comment: A method for analyzing the crystal structure of powdered materials by measuring the diffraction patterns produced when X-rays interact with randomly oriented crystallites within the sample.
alternative label: XRPD
EMMO 967080e5 2f42 4eb2 a3a9 c58143e835f9: A method for analyzing the crystal structure of powdered materials by measuring the diffraction patterns produced when X-rays interact with randomly oriented crystallites within the sample.
EMMO c84c6752 6d64 48cc 9500 e54a3c34898d: https://en.wikipedia.org/wiki/Powder_diffraction

XrdGrazingIncidence
type: Class
subclass of: ScatteringAndDiffraction